 Original research
 Open Access
 Published:
Assessment of populationbased input functions for Patlak imaging of whole body dynamic ^{18}FFDG PET
EJNMMI Physics volume 7, Article number: 67 (2020)
Abstract
Background
Arterial blood sampling is the gold standard method to obtain the arterial input function (AIF) for quantification of whole body (WB) dynamic ^{18}FFDG PET imaging. However, this procedure is invasive and not typically available in clinical environments. As an alternative, we compared AIFs to populationbased input functions (PBIFs) using two normalization methods: area under the curve (AUC) and extrapolated initial plasma concentration (C_{P}*(0)). To scale the PBIFs, we tested two methods: (1) the AUC of the imagederived input function (IDIF) and (2) the estimated C_{P}*(0). The aim of this study was to validate IDIF and PBIF for FDG oncological WB PET studies by comparing to the gold standard arterial blood sampling.
Methods
The Feng ^{18}FFDG plasma concentration model was applied to estimate AIF parameters (n = 23). AIF normalization used either AUC(0–60 min) or C_{P}*(0), estimated from an exponential fit. C_{P}*(0) is also described as the ratio of the injected dose (ID) to initial distribution volume (iDV). iDV was modeled using the subject height and weight, with coefficients that were estimated in 23 subjects. In 12 oncological patients, we computed IDIF (from the aorta) and PBIFs with scaling by the AUC of the IDIF from 4 time windows (15–45, 30–60, 45–75, 60–90 min) (PBIF_{AUC}) and estimated C_{P}*(0) (PBIF_{iDV}). The IDIF and PBIFs were compared with the gold standard AIF, using AUC values and Patlak K_{i} values.
Results
The IDIF underestimated the AIF at early times and overestimated it at later times. Thus, based on the AUC and K_{i} comparison, 30–60 min was the most accurate time window for PBIF_{AUC}; later time windows for scaling underestimated K_{i} (− 6 ± 8 to − 13 ± 9%). Correlations of AUC between AIF and IDIF, PBIF_{AUC(30–60)}, and PBIF_{iDV} were 0.91, 0.94, and 0.90, respectively. The bias of K_{i} was − 9 ± 10%, − 1 ± 8%, and 3 ± 9%, respectively.
Conclusions
Both PBIF scaling methods provided good mean performance with moderate variation. Improved performance can be obtained by refining IDIF methods and by evaluating PBIFs with testretest data.
Background
A whole body (WB) dynamic PET acquisition enables ^{18}FFDG parametric imaging. Full kinetic modeling analysis of ^{18}FFDG using WB dynamic PET requires tissue timeactivity curves (TACs) measured by PET and the arterial input function (AIF). The Patlak plot model [1, 2] can then be applied to these data to compute the net influx parameter, K_{i}, which is proportional to the glucose metabolic rate.
The AIF is obtained by collecting arterial blood samples and measuring the radioactivity concentration in the arterial plasma; these data are generally considered to be the gold standard. This invasive measurement can be associated with patient discomfort and additional exposure to personnel. Additionally, serial arterial blood sampling is not typically feasible in a clinical environment. Therefore, an alternative to arterial blood sampling for estimating the input function (IF) is desired for routine use. Several alternative methods have been proposed to replace the AIF: arterialized venous blood sampling [3], imagederived input function (IDIF) estimation [4,5,6], and populationbased input function (PBIF) modeling [7,8,9,10]. Venous blood sampling is more convenient than arterial blood sampling, but it is still invasive, especially with arterialization, i.e., sampling blood from a hand immersed in 44 °C water [11]. Heating the hand causes a vascular dilatation and increases the blood flow to the hand, so that venous samples are similar to arterial samples [12].
Measures of blood activity can be obtained by WB PET scans that typically cover large arterial blood regions such as the left ventricle and aorta; however, the accuracy of IDIFs will be affected by body motion and partial volume effects. Furthermore, the injection must be performed with the patient on the bed in order to measure the early phase of the IDIF, further compromising a clinically established workflow. The PBIF method starts with the generation of a normalized average of measured arterial blood data from several subjects (template PBIF). The PBIF method assumes that the shape of the IFs of all subjects is the same. This assumption may be violated in some patients if tracer absorption differs. The PBIF method also requires the determination of an appropriate factor to scale the template PBIF for each patient, which is another possible source of error.
In this paper, we applied both IDIF and PBIF methods to ^{18}FFDG WB PET data of oncologic patients and compared the performance of these methods with the gold standard of arterial blood sampling denoted as AIF in this paper, by assessing the Patlak K_{i} values. To generate the template PBIF, we applied two normalization methods. These template PBIFs were normalized for each subject using several scaling factors: (1) a scaling factor consisting of injected dose (ID) and initial distribution volume (iDV) of ^{18}FFDG [10] and (2) the area under the curve (AUC) of the IDIF using several time windows. While there has been substantial literature over many years developing IDIFs and PBIFs, this paper has a number of unique characteristics: (1) use of a modern PET system to extract IDIF and assess tumor quantification, (2) comparison to gold standard arterial samples, (3) use of commercial algorithms to define the aorta region of interest (ROI), and (4) comprehensive evaluation of scaling methods for the PBIF.
Material and methods
The abbreviations are listed in Table 1.
Human subjects and PET scan procedure
A total of 35 subjects were recruited for this study (Table 2). All subjects provided written consent. The study was performed in accordance with the ethical standards as laid down in the 1964 Declaration of Helsinki and federal guidelines and regulations of the USA for the protection of human research subjects contained in Title 45 Part 46 of the Code of Federal Regulations (45 CFR 46).
The subjects were divided into 2 groups: a PBIF generation group (n = 23; 11 healthy controls (HCs) and 12 clinical subjects (posttraumatic stress disorder (n = 6), epilepsy (n = 3), cocaine addiction (n = 3))) and a PBIF validation group (n = 12; oncologic subjects). In the validation group, tumors or hypermetabolic nodes were located in palate, neck, thyroid, esophagus, axilla, lung, mediastinum, inguen, and femoral shaft.
^{18}FFDG was injected by pump using a 1min infusion and arterial blood sampling was performed for 90 min in all subjects except for 1 subject (60 min). Discrete blood samples were manually drawn every 10 s from 0 to 90 s, every 15 s from 90 s to 3 min, and then at 4, 5, 6, 8, 10, 15, 20, 25, 30, 45, 60, 75, and 90 min postinjection. Samples were centrifuged to obtain plasma and then counted with a crosscalibrated well counter to produce the AIF in units of Bq/mL decay corrected to injection time.
PET scans were acquired for 90 min on a 4ring Biograph mCT PET/CT scanner concurrently with arterial blood sampling for the PBIF validation group (n = 12). A single bed cardiac PET scan was acquired for the first 6 min, followed by continuous bed motion dynamic whole body scans (2 min × 4 passes, 5 min × 15 passes). The subjects were scanned from top of the head to the knee. The dynamic data were reconstructed using OSEM (2 iterations, 21 subsets) using point spread function recovery and time of flight information, with a matrix size of 400 × 400 and 5 mm full width at half maximum Gaussian postreconstruction filtering. The data were corrected for attenuation, randoms, and scatter, but not for motion. The CT scan was not coregistered to PET since it was acquired immediately before the ^{18}FFDG injection. However the quality of the alignment was visually checked.
Normalization of AIF
The first step to generate a template PBIF curve is to normalize the amplitude of each AIF. The AIFs from the PBIF generation group were normalized in two ways. The first method used the AUC from 0 to 60 min of the AIF. For the PBIF generation group, each AIF was divided by its AUC. The second method was to use the method proposed by Vriens et al. [10], denoted as the iDV (initial distribution volume) method. The AIFs were normalized with the extrapolated initial plasma concentration of ^{18}FFDG (C_{P}*(0)). C_{P}*(0) is the expected plasma concentration under the assumption of instantaneous mixing of ^{18}FFDG at t = 0 [13]. C_{P}*(0) was obtained by fitting a portion of the curve (5 ≤ t ≤ 30 min) with an exponential function (C_{P}*(t) = C_{P}*(0)exp(αt)) [14]. Each AIF was divided by its estimated C_{p}*(0).
The iDV is the ratio of the injected dose (ID) to the initial FDG concentration, C_{P}*(0 )[14] and is effectively the volume of blood that accounts for the early distribution of tracer throughout the body. The value of iDV can be approximated noninvasively using the subject body weight and height as follows:
where c, h, and w are predetermined coefficients. These three coefficients were estimated from the individual values of iDV (=ID/C_{P}*(0)), height, and weight of the subjects in the PBIF generation group. Specifically the coefficients h and w were first determined by minimizing the coefficient of variation of c (COV_{c}) [8, 10]. Then, the coefficient, c, was determined as the mean of iDV/[(height)^{h}(weight)^{w}] among subjects.
Creation of PBIF
In the next step to generate a template PBIF curve, the normalized AIF (by AUC and iDV methods) was modeled using a compartment model that describes tracer behavior in the circulatory system proposed by Feng et al. [9].
where λ_{1}, λ_{2}, and λ_{3} are the eigenvalues of the model; A_{1}, A_{2}, and A_{3} are the coefficients; and τ is the delay constant.
Since Feng’s model describes the plasma as an impulse response function, i.e., from a true bolus injection, the model was convolved with a rectangular function (f(t) = 1, 0 ≤ t ≤ 1; f(t) =0, otherwise) to take into account our injection protocol (1min bolus). Feng’s model was applied twice. First, nonlinear least square fitting was applied to obtain the 7 parameters for each subject of the PBIF generation group. Each modelfitted normalized AIF was corrected for its estimated delay (τ) and then averaged. Next, Feng’s model was again applied to the average curve to obtain a final parameter set. The fitted PBIFs using both normalization methods are thereafter denoted as PBIF_{AUC} and PBIF_{iDV}. In the PBIF generation group, the shapes of two PBIFs were compared as follows. First, the parameters (λ_{1}, λ_{2}, λ_{3}) and the ratios of scale parameters (A_{2}/A_{1}, A_{3}/A_{1}) were compared between PBIF_{AUC} and PBIF_{iDV}. Next, the Patlak K_{i} values were compared using PBIF_{AUC} and PBIF_{iDV} that were scaled to have the same AUC.
IDIF
In the validation group, an IDIF was generated from descending aorta region automatically defined on the CT, which was used for PET attenuation correction, by a cylindrical ROI using the vendor’s ALPHA technology. The organ region of interest prediction was conducted using a learningbased algorithm [15] for automatic medical image annotation. Multiple focal anatomical structures were detected by a learningbyexample landmark detection algorithm and then inconsistent findings were eliminated through a robust sparse spatial configuration algorithm.
Subject scaling of PBIF for validation
The template PBIFs must be scaled for each individual subject, and the scaled PBIF is denoted as sPBIF. For PBIF_{AUC}, the scaling factor was determined based on the tail part of IDIF (from 15 to 90 min postinjection) using 4 different time windows. The length of the time window for scaling was 30 min, i.e., the same as the length for Patlak plot computation (see below). Multiple time windows were used as it was likely that effects such as motion and partial volume effects would produce differences in bias. Four different time windows (15–45, 30–60, 45–75, and 60–90 min) were used to scale the template PBIFs by multiplication by the AUC of the IDIF in each window (sPBIF_{AUC(15–45)}, sPBIF_{AUC(30–60)}, sPBIF_{AUC(45–75)}, sPBIF_{AUC(60–90)}). For PBIF_{iDV}, the scaling factor was computed using the injected dose and the estimated iDV using each subject’s weight and height with Eq. 1. To evaluate the robustness of iDV estimates, iDV was estimated in 3 ways, using the coefficients c, w, and h from this study, and also with the coefficients from 2 previous studies [8, 10]. In addition, to evaluate the results that could be obtained with the “best possible” scaling factor (i.e., using the subject’s plasma data), we also computed the ratio of the measured plasma to PBIF_{iDV} at 4 time points (30, 45, 60, and 75 min postinjection) for each subject. The average of these 4 ratios was used as a scaling factor to obtain sPBIF_{PLAS}.
In total, 9 estimated IFs (1 IDIF, 3 sPBIF_{iDV}, 1 sPBIF_{PLAS}, and 4 sPBIF_{AUC}) were obtained per scan for validation.
Comparison of the scaled PBIFs with IDIF and AIF
The performance of the 9 estimated IFs was compared in the validation group using the AIF as the gold standard. Two outcome measures were used to evaluate the performance: the AUC of the IF and the Patlak K_{i}. ROIs for tumors or hypermetabolic nodes were manually delineated on multiple slices of the summed (60–90 min postinjection) PET images. The size of ROI was 3.46 ± 2.21 mL (one ROI per subject). The ROIs were applied to generate timeactivity curves (TACs). The net influx rate constant (K_{i}) and the exchangeable distribution volume (V_{e}, intercept of Patlak plot) were determined for the ROI TACs using each IF and Patlak analysis applied to the period of 60–90 min postinjection. Specifically, we used a multilinear analysis to estimate K_{i} and V_{e} using the following equation:
Effect of whole blood to plasma ratio
The PBIF curves generated here were created from plasma data. However, in the above assessment, the IDIF, which measures whole blood, was not corrected for the whole blood to plasma ratio, and PBIF_{AUC} was scaled using the AUC of the uncorrected IDIF. In a separate analysis, we assessed the effect of the difference between concentrations of ^{18}FFDG in whole blood and plasma by determining the resulting bias in K_{i}. The whole blood to plasma ratio was computed from 40 s to 90 min postinjection in the PBIF validation group.
Statistical analysis
Correlations between the AUC and K_{i} with the estimated IFs and the AIF were assessed by Pearson r, mean bias, and standard deviation (SD) of bias. Statistical analysis was performed by Prism 8 (GraphPad Software). All kinetic modeling was performed with inhouse programs written with IDL 8.0 (ITT Visual Information Solutions, Boulder, CO).
Results
Creation of PBIF
The parameters from fitting of the AIFs by Feng’s model using the AUC and C_{P}*(0) (ID/iDV) normalizations are summarized in Table 3. The shaperelated parameters (λ_{1}, λ_{2}, λ_{3}) were very similar between PBIF_{AUC} and PBIF_{iDV.} The values of the relative amplitudes A_{2}/A_{1} and A_{3}/A_{1} were similar between the two PBIFs: 0.010 and 0.008 for PBIF_{AUC} and 0.009 and 0.006 for PBIF_{iDV}, respectively.
To compare the two PBIFs, tests were performed with the two PBIFs scaled to have the same AUC. In that case, Patlak K_{i} values using PBIF_{AUC} were almost identical to those using PBIF_{iDV} (K_{i}(PBIF_{AUC}) = 0.994 × K_{i}(PBIF_{iDV}) − 0.002, R^{2} = 1.000), indicating that there is no meaningful difference between the shapes of the two PBIFs.
Comparing the contribution of the terms of Eq. 2 to the PBIF, the third term (\( {A}_3{e}^{{\lambda}_3\left(t\tau \right)} \)) accounted for > 95% of the PBIF after 16 min postinjection.
IDIF
In the validation group, the volume of the aorta ROI was 1.55 ± 0.11 mL. Figure 1a shows a comparison of a typical IDIF and its corresponding AIF. The IDIF tends to undershoot the AIF at early times (t < 20 min) and overshoot it at late times (t > 30 min), with varying degree of under/overshoot among subjects (%difference, − 7% ± 8% (t < 20 min) and 13% ± 12% (t > 30 min)). Fitting Feng’s model to IDIFs and AIFs, the third eigenvalue λ_{3} of the IDIF was significantly smaller than that of AIF (IDIF, 0.008 ± 0.002 min^{−1}and AIF, 0.011 ± 0.002 min^{−1}, P = 0.008).
Subject scaling of PBIF for validation
The median iDV was 13.1 L (mean ± SD = 13.0 ± 1.7), which corresponds to 0.14 L/kg body weight. Table 4 shows the three estimated coefficients (c, h, w) in our study (from the PBIF generation group) compared to previous references. Those coefficients were used to predict C_{P}*(0) and compare to the actual values from blood samples in the validation group. Using values in this study, differences were acceptable (3 ± 8%). For the literature values, although the coefficients themselves were quite different, the percent bias of the estimated C_{P}*(0) was reasonable, especially for the values from Vriens et al. [10].
Comparison of the scaled PBIFs with IDIF and AIF
In the validation group, comparisons between AUC(0–90 min) and Patlak K_{i} with respect to the AIF values are shown in Tables 5 and 6, respectively.
For AUC, the early time windows, 15–45 min or 30–60 min, for scaling PBIF_{AUC} provided similarly good performance (0–90 min) in terms of Pearson r, bias, and SD (Table 5). Later time windows produced poorer correlation and overestimated the AUC(0–90 min). Typical sPBIFs are shown in Fig. 2 where the differences in scaling are best visualized in the tail of the curve. The correlation, bias, and SD were similar between IDIF, sPBIF_{AUC} with the best time window, sPBIF_{iDV}, and sPBIF_{PLAS} (correlation, 0.90–0.94; bias, − 1 to 3%; SD, 5–6%).
Figure 3 shows individual K_{i} bias values using the IDIF or any of the sPBIFs, with K_{i} estimated using the AIF as the gold standard. The %bias was particularly large (− 47 and − 60%; Fig. 3a) for small K_{i} values (< 0.01 mL/min/cm^{3}) with the IDIF. Therefore, the K_{i} bias (Table 6) was calculated in two ways, i.e., with and without these two tumors. Unlike the IDIF method, the K_{i} bias using all PBIF values was not affected by the magnitude of K_{i} (Fig. 3b, c).
When AUC was overestimated, K_{i} was generally underestimated (Table 6). Patlak K_{i} determined by the IDIF was lower than the gold standard values (using the AIF) (− 9%), although the correlation was similar to those of other PBIFs (0.99–1.00). For sPBIF_{AUC}, K_{i} was underestimated when using late time windows to scale the PBIF_{AUC} (− 14% using 60–90 min). Conversely, using early time windows for scaling, the correlation, bias, and SD of sPBIF_{AUC} was closest to those of sPBIF_{PLAS}, which represents the bestpossible outcome. For sPBIF_{iDV}, using scaling coefficients from this study, the mean bias was low, the SD of the bias was similar to other methods, and the correlation lower than with sPBIF_{AUC}. Using scaling coefficients from other published studies for sPBIF_{iDV} led to larger mean bias and similar correlation and SD.
Effect of whole blood to plasma ratio
The whole blood to plasma ratio increased from a mean of 0.93 to 0.97 over 90 min (Fig. 4): The whole blood/plasma curve could be described by the function 0.97 − 0.06 × exp(− 0.08 × t). The mean ratio did not differ between 30 min (0.95 ± 0.05) and at 90 min postinjection (0.97 ± 0.05). The mean whole blood to plasma ratio was 0.97 ± 0.04 (15–45 min), 0.96 ± 0.03 (30–60 min), 0.97 ± 0.03 (45–75 min), 0.97 ± 0.04 (60–90 min), and 0.94 ± 0.03 (40 s–90 min). Applying the above mean whole blood to plasma ratio values for correction to the IDIF increased its value, so K_{i} values became even more underestimated: the mean bias of K_{i} became − 14% (IDIF), 0% (sPBIF_{AUC(15–45)}), − 4% (sPBIF_{AUC(30–60)}), − 9% (sPBIF_{AUC(45–75)}), and − 16% (sPBIF_{AUC(60–90)}) instead of the values in Table 6 (n = 10).
Discussion
This study compared the performance of PBIFs with different normalization and scaling methods for the purpose of measuring the Patlak uptake constant K_{i} for ^{18}FFDG. The PBIFs were compared to IDIF and AIFs, with the latter used as the gold standard.
Two forms of the PBIF were generated from arterial sample data using two normalization methods (AUC or C_{P}*(0)) and were first compared. The K_{i} values using PBIF_{AUC} were almost identical to those using PBIF_{iDV}. This suggests that the PBIF shape was not affected by the different normalization methods. Therefore, the comparison among PBIFs was reduced to the comparison of scaling factors.
To apply the PBIFs without the need for blood sampling, we tested two scaling methods. We also scaled the PBIF using the measured plasma samples for each scan to define the best achievable results by PBIF. Four plasma samples at 30, 45, 60, and 75 min postinjection were used for scaling to reduce effects of measurement noise in the plasma. The sPBIF_{PLAS} overestimated K_{i} by 2 ± 6 %, due to slight differences in IF shape between subjects. Thus, ideally, a bloodfree PBIF method could achieve comparable results.
One scaling method used a part of the IDIF. In WB PET imaging, large blood pools are always available. As shown in Fig. 1, the estimated IDIF showed a consistent pattern compared to the AIF, with undershoot at early times and overshoot at late times, perhaps due to partial volume averaging, but the magnitude of under/overshoot was different among subjects. Therefore, the Patlak K_{i} was significantly underestimated using the PBIFs scaled by the late AUC values from the IDIF. The best time window for scaling (in terms of minimum bias) was 30–60 min (bias, − 1% and SD, 8%; Table 6). In that case, however, the required scan time would be 1 h, 30–60 min to measure the part of the IDIF used for scaling, and 60–90 min for Patlak K_{i}. Note that the SD of bias was very similar for all sPBIF_{AUC} time periods; thus, if a mean bias was acceptable, e.g., if that bias was consistent across scans in the same patient, then later time periods could be used for scaling, providing a short scan.
The second scaling method used the estimated C_{p}*(0), the extrapolated initial ^{18}FFDG plasma concentration. This scaling approach has potential advantages since it does not require the IDIF for scaling and thus has a short scan and is not subject to effects of body motion and partial volume effect on the IDIF. Vriens et al. [10] reported a median iDV of 0.168 L/kg, slightly higher than the value in our study (0.144 L/kg). We fitted the iDV equation (Eq. 1) using the same method as Shiozaki et al. and Vriens et al. and found quite different values for the estimated coefficients (c, h, w). The estimated C_{P}*(0) values using the injected dose and these coefficients were compared with the extrapolated C_{P}*(0) values measured from the AIF. Not surprisingly, the bias of C_{P}*(0) was smallest using our fitted parameters. The coefficient estimation might be affected by the study population or other methodological details. For example, the difference in body habitus of the study subjects at different sites might affect the results. Also, the estimation is affected by the correlation between height and weight which introduces instability in the parameters h and w. Patlak K_{i} estimated with this PBIF scaling method produced minimal bias and similar SD to the other scaling methods.
The mean biases of AUC(0–90 min) using IDIF, sPBIF_{AUC} with early time windows, and sPBIF_{iDV} were all minimal. However, a large negative mean bias of K_{i} with the IDIF was found, which was much larger than the other PBIF methods. Specifically, K_{i} with the IDIF was greatly underestimated (as a percentage) for small K_{i} values, while this was not observed for K_{i} with PBIF (Fig. 3). This difference in the K_{i} bias is due to the differences in the shapes of the IDIF and the AIF. The input function parameter λ_{3} (the terminal clearance rate) of the IDIF was much smaller than that of the AIF or the PBIFs, i.e., the IDIF showed slower clearance than the other IFs, resulting in large % underestimation of K_{i} for small K_{i} values.
To clarify this finding, we performed a simulation to assess the effect of λ_{3} on K_{i} estimates for large and small K_{i} values. Three IFs were computed using different λ_{3} values (0.012, 0.0084, 0.0048 min^{−1}) (Figure S1A) with all normalized to have the same AUC. Two TACs were computed using the input function with λ_{3} = 0.012 (Figure S1B) having different K_{i} values (0.0077, 0.077 mL/min/cm^{3}) but the same V_{e} (0.42). The Patlak plot was computed for these two TACs using three IFs, i.e., the correct IF and the two with slower terminal clearance (Figure S1C and D). As shown in Table S1, K_{i} was underestimated, with much larger percent bias for small K_{i} values using the IFs with small λ_{3} values. The underestimated K_{i} was compensated by an overestimated intercept value, which has a larger error for larger K_{i}.
In several past reports [10, 16], the IDIF, which measures whole blood, was used as IF without correction for the difference between concentrations of ^{18}FFDG in whole blood and plasma, assuming these differences are small [17]. In our study, we also used the uncorrected IDIF for Patlak analysis (Table 6). To assess this effect, the whole blood to plasma ratio was computed. Mean whole blood to plasma ratio increased monotonically from 0.93 to 0.97 over 90 min (i.e., the mean plasma to whole blood ratio decreased from 1.09 to 1.03). Similar results were reported previously (1.09 to 1.04 [11] and 1.12 to 1.07 [18] over 90 min). When the whole blood to plasma ratio is taken into consideration, mean underestimation of K_{i} by the IDIF method worsened slightly.
Several ^{18}FFDG tumor imaging guidelines reviewed in [19] suggested that a static scan should start at 30~40 min or 50~70 min postinjection, but an ideal time window (length and starting time) for tumor Patlak analysis is not clearly defined. In a brain study using healthy subjects, Lucignani et al. [20] reported that Patlak K_{i} is stable using a 30min window in the interval between 45 and 120 min postinjection. In our study, we used a 60–90min time window for Patlak analysis; this time period can also be used to generate a static SUV image by appropriate image averaging.
Comparing the results of our scaled PBIF methods, sPBIF_{AUC(30–60)} and sPBIF_{iDV} produced similarly small bias and high correlation coefficients in Patlak K_{i} estimation. In the PBIF_{AUC} method, no bias will be introduced due to an inaccurate dose calibrator crosscalibration to the PET scanner; however, errors in this calibration affect the PBIF_{iDV} method. PBIF_{AUC(30–60)} requires a 1h scan when the Patlak time window is set from 60 to 90 min, while the PBIF_{iDV} requires scan time for the Patlak analysis only. Also, measurement of body weight, height, and injected dose is simpler than obtaining IDIF curves, depending on the available tools in each clinical environment. Therefore, PBIF_{iDV} would provide a simple protocol than PBIF_{AUC(30–60)}. Using the methodology shown here, both approaches showed acceptable performance. sPBIF_{AUC} has slightly better performance, but sPBIF_{iDV} should be easier to implement in clinical setting, although some sitespecific tuning of the iDV coefficients may be necessary.
In addition to considering mean bias, the SD of bias (~ 9%) for all sPBIF methods was larger than the best possible attainable value using the subject’s own plasma data (sPBIF_{PLAS}, 6%). Since variances add in quadrature, this difference in SD suggests that an additional error of 6–7% is introduced by the IDIF AUC and iDV scaling methods. While it is not clear how to improve the iDV scaling method, IDIF performance would likely be improved by changing the shape of the ROI, as well as applying motion correction and partial volume correction. Since the IDIF ROI was defined from the CT, we assessed the effects of misalignment between the CT and PET on the AUC of the IDIF. The IDIF ROI was shifted by 1 to 6 voxels (i.e., 2 to 12 mm) in the x (leftright), y (anteriorposterior), and z (superiorinferior) directions, and we determined the maximum misalignment in each direction leading to ≤ 5% decrease in the AUC (15–45, 30–60, 45–75, and 60–90 min) from the shifted ROI. The most sensitive directions to misalignment were y (5 to 7 mm) and x (6 to 11 mm); the z direction showed minimal effects, as expected. The earlier time window was more sensitive to misalignment due to the higher contrast between the aorta and background. Partial volume effects would be a major contributing factor to the overestimation of AUC, especially in later time windows, as seen in Table 5 (19% overestimation of AUC(0–90 min) using sPBIF_{AUC(60–90)}). If the quality of the IDIF ROI is improved, e.g., with motion and partial volume corrections, so that the later part of the IDIF can provide an accurate value, then the bias of K_{i} using PBIF_{AUC(60–90)} would be improved. In particular, in a typical clinical protocol, where the PET scan begins at 60 min, there will be less delay between CT and PET scans, so motion issues would likely be reduced. Also, we believe that using the imaging data to directly quantify the IF is of value, since daytoday variation in the IF cannot be captured by the iDV method.
As described above, we assessed relative performance of the methods by calculating accuracy (mean % bias) and variability (SD of % bias). Both of these measures are relevant, although the relative importance depends on the clinical question. A small mean bias compared to the AIF means that the method is intrinsically accurate over the entire patient group. However, the SD of the bias across subjects and tumors should also be considered. If the SD is large, then the ability to reliably measure changes in tracer uptake between scans of the same patient may be poor. Alternatively, if large SD across patients is caused by subjectspecific biases, e.g., due to IDIF ROI definition (excluding motion effects), which remain consistent across scans, then such variability may be clinically acceptable if the goal is to assess treatment response. Thus, the best way to fully assess the performance of PBIFs would be with testretest data using the reproducibility of the estimated K_{i} as the key outcome measure.
Recent improved detector technology and clinical application demands led to the development of total body PET systems [21, 22], such as the uEXPLORER [23, 24] and PennPET Explorer [25]. Access for arterial blood sampling site is challenging in these systems. However, since the aorta is always in the field of view and the acquired dynamic data will have lower noise, the PBIF methods will be useful and compatible with these total body PET scan systems.
Conclusions
In this paper, using a modern PET system, we assessed and optimized IDIFs and PBIFs using arterial blood samples and commercial software to define the IDIF ROI. We applied these IDIF and PBIF methods for FDG oncological WB PET studies. The PBIF methods scaled by either IDIF AUC or ID and iDV showed good performance, with a small mean bias and moderate variability, whereas the IDIF method produced negative mean bias of K_{i}. Further improvements in accuracy and precision can be obtained with motion correction and partial volume corrections.
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the corresponding author on reasonable request.
References
Patlak CS, Blasberg RG, Fenstermacher JD. Graphical evaluation of bloodtobrain transfer constants from multipletime uptake data. J Cereb Blood Flow Metab. 1983;3:1–7.
Karakatsanis NA, Zhou Y, Lodge MA, Casey ME, Wahl RL, Zaidi H, et al. Generalized wholebody Patlak parametric imaging for enhanced quantification in clinical PET. Phys Med Biol. 2015;60:8643–73.
van der Weerdt AP, Klein LJ, Visser CA, Visser FC, Lammertsma AA. Use of arterialised venous instead of arterial blood for measurement of myocardial glucose metabolism during euglycaemichyperinsulinaemic clamping. Eur J Nucl Med Mol Imaging. 2002;29:663–9.
Chen K, Bandy D, Reiman E, Huang SC, Lawson M, Feng D, et al. Noninvasive quantification of the cerebral metabolic rate for glucose using positron emission tomography, 18Ffluoro2deoxyglucose, the Patlak method, and an imagederived input function. J Cereb Blood Flow Metab. 1998;18:716–23.
Asselin MC, Cunningham VJ, Amano S, Gunn RN, Nahmias C. Parametrically defined cerebral blood vessels as noninvasive blood input functions for brain PET studies. Phys Med Biol. 2004;49:1033–54.
van der Weerdt AP, Klein LJ, Boellaard R, Visser CA, Visser FC, Lammertsma AA. Imagederived input functions for determination of MRGlu in cardiac (18)FFDG PET scans. J Nucl Med. 2001;42:1622–9.
Takikawa S, Dhawan V, Spetsieris P, Robeson W, Chaly T, Dahl R, et al. Noninvasive quantitative fluorodeoxyglucose PET studies with an estimated input function derived from a populationbased arterial blood curve. Radiology. 1993;188:131–6.
Shiozaki T, Sadato N, Senda M, Ishii K, Tsuchida T, Yonekura Y, et al. Noninvasive estimation of FDG input function for quantification of cerebral metabolic rate of glucose: optimization and multicenter evaluation. J Nucl Med. 2000;41:1612–8.
Feng D, Huang SC, Wang X. Models for computer simulation studies of input functions for tracer kinetic modeling with positron emission tomography. Int J Biomed Comput. 1993;32:95–110.
Vriens D, de GeusOei LF, Oyen WJ, Visser EP. A curvefitting approach to estimate the arterial plasma input function for the assessment of glucose metabolic rate and response to treatment. J Nucl Med. 2009;50:1933–9.
Phelps ME, Huang SC, Hoffman EJ, Selin C, Sokoloff L, Kuhl DE. Tomographic measurement of local cerebral glucose metabolic rate in humans with (F18)2fluoro2deoxyDglucose: validation of method. Ann Neurol. 1979;6:371–88.
Goldschmidt SI, Light AB. Method of obtaining from veins blood similar to arterial blood in gaseous content. J Biol Chem. 1925;64:53–8.
Sadato N, Tsuchida T, Nakaumra S, Waki A, Uematsu H, Takahashi N, et al. Noninvasive estimation of the net influx constant using the standardized uptake value for quantification of FDG uptake of tumours. Eur J Nucl Med. 1998;25:559–64.
Goodman LS, Gilman A, Brunton LL, HilalDandan R, Knollmann BC. Goodman & Gilman's the pharmacological basis of therapeutics. New York [etc.]: McGraw Hill Education; 2018.
Tao Y, Peng Z, Krishnan A, Zhou XS. Robust learningbased parsing and annotation of medical radiographs. IEEE Trans Med Imaging. 2011;30:338–50.
ZanottiFregonara P, Maroy R, Comtat C, Jan S, Gaura V, BarHen A, et al. Comparison of 3 methods of automated internal carotid segmentation in human brain PET studies: application to the estimation of arterial input function. J Nucl Med. 2009;50:461–7.
Gambhir SS, Schwaiger M, Huang SC, Krivokapich J, Schelbert HR, Nienaber CA, et al. Simple noninvasive quantification method for measuring myocardial glucose utilization in humans employing positron emission tomography and fluorine18 deoxyglucose. J Nucl Med. 1989;30:359–66.
Gheysens O, Postnov A, Deroose CM, Vandermeulen C, de Hoon J, Declercq R, et al. Quantification, variability, and reproducibility of basal skeletal muscle glucose uptake in healthy humans using 18FFDG PET/CT. J Nucl Med. 2015;56:1520–6.
Thie JA, Hubner KF, Smith GT. Optimizing imaging time for improved performance in oncology PET studies. Mol Imaging Biol. 2002;4:238–44.
Lucignani G, Schmidt KC, Moresco RM, Striano G, Colombo F, Sokoloff L, et al. Measurement of regional cerebral glucose utilization with fluorine18FDG and PET in heterogeneous tissues: theoretical considerations and practical procedure. J Nucl Med. 1993;34:360–9.
Vandenberghe S, Moskal P, Karp JS. State of the art in total body PET. EJNMMI Phys. 2020;7:35.
Surti S, Pantel AR, Karp JS. Total Body PET: Why, how, what for? Ieee T Radiat Plasma. 2020;4:283–92.
Badawi RD, Shi H, Hu P, Chen S, Xu T, Price PM, et al. First human imaging studies with the EXPLORER totalbody PET scanner. J Nucl Med. 2019;60:299–303.
Zhang X, Cherry SR, Xie Z, Shi H, Badawi RD, Qi J. Subsecond totalbody imaging using ultrasensitive positron emission tomography. Proc Natl Acad Sci U S A. 2020;117:2265–7.
Pantel AR, Viswanath V, DaubeWitherspoon ME, Dubroff JG, Muehllehner G, Parma MJ, et al. PennPET Explorer: human imaging on a wholebody imager. J Nucl Med. 2020;61:144–51.
Acknowledgements
The authors appreciate the excellent technical assistance of the staff at the Yale University PET Center. This work was made possible by NIH grant 1S10RR029245 and CTSA grant number UL1 TR000142 from the National Center for Advancing Translational Sciences (NCATS), a component of the National Institutes of Health (NIH). Its contents are solely the responsibility of the authors and do not necessarily represent the official view of NIH.
Copyright statement
The author CY is a military service members or federal employee. This work was prepared as part of the authors’ official duties. Title 17 U.S.C 105 provides that “Copyright protection under this title is not available for any work of the United States Government.” Title 17 U.S.C. 101 defines a US Government work as a work prepared by a member or employee of the US Government as a part of that person’s official duties.
Disclaimer

1.
The identification of specific products or scientific instrumentation is considered an integral part of the scientific endeavor and does not constitute endorsement or implied endorsement on the part of the author, DoD, or any component agency. The views expressed in this manuscript are those of the author and do not reflect the official policy of the Department of Army/Navy/Air Force, Department of Defense, or US Government.

2.
We certify that all individuals who qualify have been listed; that each has participated in the conception and design of this work, the analysis of data, the writing of the document, and the approval submission of this version; that the document represents valid work; and that each takes public responsibility for it.
Funding
This study was funded by Siemens.
Author information
Authors and Affiliations
Contributions
MN contributed to the data analysis, interpretation of data, and manuscript preparation. JDG and VS contributed to methodology of the study and interpretation of data. TM contributed to investigation of the study and wrote programs used in the work. CY and MD contributed to subject recruitment and care. MKC contributed to subject evaluation and provided clinical diagnosis. AS and RC provided overall supervision of the study design and execution. MN, JDG, VS, TM, MKC, CY, AS, and RC joined in the discussions and editing the manuscript. All authors read and approved the final manuscript.
Corresponding author
Ethics declarations
Ethics approval and consent to participate
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki Declaration and its later amendments or comparable ethical standards. This study was approved by the Yale University Human Investigation Committee and the YaleNew Haven Hospital Radiation Safety Committee.
Informed consent was obtained from all patients included in this study.
Consent for publication
Not applicable
Competing interests
VS and AS are employees of Siemens Healthineers.
Additional information
Publisher’s Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Supplementary information
Additional file 1: Figure S1.
(A) Three input functions simulated with different λ_{3} values (0%, 30%, 60% lower than mean value of 0.012 min^{1}) and the same area under the curve. Dotted curves show the difference from the input function with λ_{3}=0.012; (B) two timeactivity curves (TACs) computed using the input function (λ_{3}=0.012). These curves have different K_{i} and the same V_{e} values, as specified in the legend; (C) Patlak plots of the TAC with the low K_{i} using the three input functions; (D) Patlak plots of the TAC with the high K_{i} using three input functions. Note the difference in yaxis scaling of (C) and (D). Table S1. Effect of λ_{3} of input function on the K_{i} estimation
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to theoriginal author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made. The images orother third party material in this article are included in the article's Creative Commons licence, unless indicated otherwise in a creditline to the material. If material is not included in the article's Creative Commons licence and your intended use is not permitted bystatutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view acopy of this licence, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Naganawa, M., Gallezot, JD., Shah, V. et al. Assessment of populationbased input functions for Patlak imaging of whole body dynamic ^{18}FFDG PET. EJNMMI Phys 7, 67 (2020). https://doi.org/10.1186/s4065802000330x
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/s4065802000330x
Keywords
 ^{18}FFDG
 Populationbased input function
 Whole body PET imaging
 Patlak plot