 Original research
 Open Access
 Published:
Tumor dosimetry using ^{177}Lu: influence of background activity, measurement method and reconstruction algorithm
EJNMMI Physics volume 10, Article number: 39 (2023)
Abstract
Background
Imagebased tumor dosimetry after radionuclide therapy, using the isotope ^{177}Lu, finds application e.g., for tumortoorgan dose comparison and for dose response evaluation. When the tumor extent is not much larger than the image resolution, and when ^{177}Lu is found in nearby organs or other tumors, an accurate determination of tumor dose is particularly challenging. Here a quantitative evaluation of three different methods for determining the ^{177}Lu activity concentration in a phantom is performed, and the dependence on a variety of parameters is described. The phantom (NEMA IEC body phantom) has spheres of different size in a background volume, and spheretobackground ^{177}Lu activity concentration ratios of infinity, 9.5, 5.0 and 2.7 are applied. The methods are simple to implement and wellknown from the literature. They are based on (1) a large VOI encompassing the whole sphere, without background activity and with volume information from other sources, (2) a small VOI located in the sphere center, and (3) a VOI consisting of voxels with voxel value above a certain percentage of the maximum voxel value.
Results
The determined activity concentration varies significantly with sphere size, spheretobackground ratio, SPECT reconstruction method and method for determining the concentration. Based on the phantom study, criteria are identified under which the activity concentration can be determined with a maximal error of 40% even in the presence of background activity.
Conclusions
Tumor dosimetry is feasible in the presence of background activity using the abovementioned methods, provided appropriate SPECT reconstructions are applied and tumors are selected for dosimetry analysis according to the following criteria for the three methods: (1) solitary tumor with diameter > 15 mm, (2) tumor diameter > 30 mm and tumortobackground ratio > 2, and (3) tumor diameter > 30 mm and tumortobackground ratio > 3.
Background
In radionuclide therapy using the isotope ^{177}Lu, e.g., peptidereceptor radionuclide therapy (PRRT) of neuroendocrine tumors [1, 2] using [^{177}Lu]LuDOTATOC or [^{177}Lu]LuDOTATATE, or PSMA radioligand therapy (RLT) of metastatic prostate cancers [3] using [^{177}Lu]LuPSMA, posttherapy dosimetry of organs and tumors is feasible based on quantitative gammacamera or SPECT imaging of the gamma emissions from ^{177}Lu. Imagebased organ dosimetry, in particular of kidneys, have been reported now for several years, and to some extent also tumor dosimetry [4,5,6]. Tumor dosimetry finds application e.g., for comparison of tumor and normal organ doses and for dose response evaluations.
In tumor dosimetry, a number of pertinent challenges must be considered, e.g., that tumors are often not visible on the noncontrast CTscan of a SPECT/CT protocol, that partial volume effects (spillout) can be very significant for smaller tumors, and that ^{177}Lu in nearby organs or other tumors may significantly influence the measurement of ^{177}Lu activity concentration in tumors (spillin). For example, in PRRT, quantification of ^{177}Lu in liver metastases will be affected by physiological uptake in the liver [7], and quite generally quantification of lesions in the abdominal region will be affected by uptake in the liver, kidneys and spleen. Restriction of a tumor dosimetry analysis to wellisolated tumors severely limits the number of lesions available for analysis, and it may in practice reduce the analysis to certain classes such as bone or lymph node metastases, which can be found far from the abdominal region.
The mean tumor dose can be calculated from measurements of the ^{177}Lu activity concentration in tumors at various time points after injection. A variety of methods have been employed to determine the activity concentration based on planar scintigraphy, SPECT/CT scans, or a hybrid thereof, which address or circumvent the abovementioned inherent challenges. A volumeofinterest (VOI) defined by a threshold value has been applied together with partial volume correction [8,9,10] for tumors separated from adjacent lesions. Manual placement of a small spherical VOI within a tumor [11], or of a large VOI containing the tumor with volume information from a PET/CT scan [12, 13] circumvent the need for partial volume correction. These three methods are relatively straightforward to implement, not very timeconsuming and do not require an accurate delineation of the tumor. Other methods include manual delineation [14, 15] or an advanced softwarebased segmentation based on both planar, SPECT and CT images [16]. In a very recent proposal, lesionvolume and background dependent calibration factors were determined with the intent to apply these for correction of spillin and spillout [17], but this method has currently not been applied to patient data.
The aim of this paper is to obtain practically applicable limits on tumor size and tumortobackground ratio, to aid in the selection of tumors for dosimetry analysis based on posttherapy SPECT/CT images. A phantom study was performed to evaluate to which extent the mean ^{177}Lu concentration measured in different sized tumors is influenced by a variable background activity for a quantitative SPECT protocol. This was considered for the three methods employing (1) a large VOI around a tumor with knowledge of the volume [12, 13], (2) a small spherical VOI within the tumor [11] or (3) a thresholdgenerated VOI with partial volume correction [8]. On the basis of these phantom data, limits on tumor size and tumortobackground activity ratio are suggested, for which the accuracy is judged satisfactory to provide useful results in tumor dosimetry.
Methods
Phantom
A National Electrical Manufacturers Association (NEMA) NU2 2001 image quality phantom [18] with six fillable spheres (inner diameters 10, 13, 17, 22, 28 and 37 mm; volumes 0.52, 1.15, 2.57, 5.58, 11.49 and 26.52 ml) in a 10.2 L background volume was used for the study. The spheres were filled with a solution prepared from 2.0 GBq of noncarrier added [^{177}Lu]LuCl_{3} in an excess amount of CaDTPA (400 mg in 2 ml, i.e., > 10^{6} DTPA molecules per LuCl_{3} molecule) diluted in isotonic water to a ^{177}Lu concentration of 514 kBq/ml. The CaDTPA acted as a chelator to prevent ^{177}Lu from sticking to the phantom walls [19]. The background volume was first filled with nonradioactive water. This phantom was scanned as described below, and then ^{177}Lu in solution was added stepwise to the background volume, and the phantom was scanned with spheretobackground ratios of 9.5:1, 5.0:1 and 2.7:1. Each time ^{177}Lu was added to the background volume, the NEMA phantom was turned and shaken and then allowed to homogenize by Brownian motion in the solution for at least 8 h. In the following, these phantoms are referred to by their spheretobackground ratio as P_{∞} (nonradioactive water background), P_{9.5}, P_{5.0} and P_{2.7}, respectively.
Quantitative SPECT/CT scans
Quantitative SPECT (QSPECT) scans with voxel values representing the ^{177}Lu activity concentration were obtained using a Siemens Symbia T16 SPECT/CT scanner (Siemens Medical Solutions Inc., USA) and reconstruction using orderedsubsetexpectation–maximization (OSEM) in Siemens Flash3D. The acquisition and reconstruction protocols followed our procedure for kidney dosimetry, as described in detail in Ref. [20]. Briefly, acquisition was performed with mediumenergy collimators, an energy window at 208 keV (20% width), 128 × 128 matrix (4.8 mm pixel size), 64 views in total and time/view varied as described below. Lower energy windows enabled scatter and deadtime correction. Reconstruction was performed with 4 iterations, 8 subsets and corrections for attenuation, scatter and collimator blurring. For this QSPECT procedure, no postfilter was applied. Calibration, i.e., determination of sensitivity and deadtime constant, was performed 31 months prior to this phantom study, using ^{177}Lu (84–3188 MBq) in closed containers placed in a scatter medium (eight bags each containing 500 ml of saline). The sensitivity was 1.09 × 10^{–5} s^{−1} Bq^{−1} and the deadtime 0.5 µs for the applied SPECT system. Stability of sensitivity was verified using ^{177}Lu sources with known activity. Conversion coefficients (conversion from reconstructed voxel values to ^{177}Lu concentration) depended on the time/view and were adjusted accordingly for each scan.
Scans of P_{∞} were performed three times to evaluate reproducibility, while P_{9.5} and P_{5.0} was scanned once and P_{2.7} was scanned twice. The SPECT/CT scans were performed over a 3day period, and the time per view was adjusted to obtain a similar number of counts from the spheres in each scan, starting with 90 s/view for P_{∞} 3 h after the measurement of ^{177}Lu activity.
In the largest sphere of P_{∞}, an image artifact appeared as a 'tunnel' of low intensity through the center, which was seen in the transverse plane as a central 'valley', see Fig. 1a, presumably due to resolution recovery (pointspread function correction) applied in the reconstruction [21]. In order to perform analysis without such an artifact, we performed also smoothed quantitative reconstructions by applying a 10 mm Gaussian postfilter to the (unfiltered) QSPECT reconstruction, see Fig. 1b. This reconstruction is referred to as fQSPECT ('filtered QSPECT') in the following. Also a Bayesian reconstruction with Bowsher prior penalty factor was performed in Hermes Hybrid Recon (version 1.0.30, 'AMAP' reconstruction method, 4 most similar neighbors and Bayesian weight 0.3, 4 iterations, 16 subsets with attenuation and scatter correction and resolution recovery [21]; Hermes Medical Solutions AB, Sweden), see Fig. 1c, as this was previously shown to remove the observed artifact [21]. The Bayesian reconstruction with Bowsher prior seeks to obtain SPECT images with similar voxel values in anatomically similar voxels, i.e., voxels with Hounsfield Units (HU) close to each other in the CT scan. No further scaling was applied to the fQSPECT reconstruction, as the only difference from the standard QSPECT reconstruction was the 10 mm Gaussian postfilter, and it could be verified that the total counts in the reconstructed images were essentially identical (deviation < 1.1% for P_{∞} and ≤ 0.03% for all other phantoms). For the Bayesian reconstruction a quantitative measure of activity was simply obtained by scaling the voxel values with the ratio between the sum of voxel values in the unfiltered QSPECT reconstruction to the corresponding sum in the Bayesian reconstruction.
Analysis of quantitative SPECT scans
For each scan, the three SPECT reconstructions were analyzed in PMOD 4.006 (PMOD Technologies LLC, Switzerland) to obtain several different parameters. For P_{∞} and P_{2.7} the mean and the standard deviation of these parameters were calculated. The starting point for the analysis was six spherical VOIs (VOI_{dia}) with diameters equal to the sphere diameters and aligned manually with the respective spheres on the CT scan.
Maximum and peak values and accuracy
In each sphere the maximum voxel value and the peak value, i.e., the maximum average in a 1 cm^{3} sphere, were used as references for selecting a threshold value (see below). They were recorded within each spherical VOI (VOI_{dia}), except for the peak value in the 10 mm diameter spheres, where the VOI volume was below 1 cm^{3}. These values were compared to the actual mean in the spheres.
The accuracy of the reconstructions was evaluated for P_{∞} by comparing the known activity in each sphere to the reconstructed activity within a spherical VOI with diameter equal to the sphere diameter plus 15 mm (VOI_{dia+15}, concentric with VOI_{dia}). The 15 mm extension was chosen in order to include essentially all counts from the respective spheres without inclusion of counts from neighboring spheres. The deviation was calculated with the known activity as reference.
Contrast and recovery coefficient
The contrast for each sphere was calculated as
where C_{dia} is the mean concentration in the VOI_{dia}, and C_{bkg} is the mean concentration in a large (about 4 L) VOI in the part of the phantom without spheres.
The recovery coefficients, RC, for the various sphere sizes were determined as the ratio of the activity in the VOI_{dia} to the total activity within VOI_{dia+15} [22]. For P_{9.5}, P_{5.0} and P_{2.7}, the total activity within VOI_{dia+15} would contain activity in the background volume, and hence the total activity was calculated as the mean of decay corrected activity within the corresponding VOIs (VOI_{dia+15}) of the three SPECT/CT scans of P_{∞}.
For P_{∞}, the recovery coefficients for all three reconstructions were fitted by the function
where d is the sphere diameter [8]. Since the recovery coefficients were determined from two VOIs within the same images, the recovery coefficients should converge to unity for large spheres, as is the case for RC(d) in Eq. (2).
Mean concentration for small spherical VOIs and thresholdbased VOIs
Spherical VOIs of diameter 10, 15 and 20 mm were created concentric with the VOI_{dia} for the spheres with diameter larger than or equal to 10, 15 and 20 mm, respectively. The mean concentration within the small VOIs was recorded and compared to the actual mean in the spheres.
Thresholdbased VOIs were grown with a threshold equal to 50% and 40% of the maximum voxel value within each sphere, and also with thresholds of 60% and 50% of the peak value. The mean concentration and the volume of the VOIs were recorded. If the VOIs reached a 75 mm diameter bounding sphere, they were not evaluated further. The VOIs were generally smaller than the spheres, as shown below in the Results section, unlike in the original paper by Ilan et al. [8]. This would have made partial volume correction based directly on the VOI volume incorrect in the present work. The VOIs were instead first seen as representative volumes within the sphere, and the mean concentration in the VOIs were compared to the actual mean in the spheres. Further, to apply partial volume correction along the lines of Ilan et al. [8], the following approach was used to obtain a partial volume corrected concentration, calculated on the basis of the concentration in the VOI (C_{VOI}) and its volume (V_{VOI}), see Fig. 2. The true concentration in the sphere is equal to the reconstructed activity within the sphere, A, divided by its volume V and the recovery coefficient RC(d), see Eq. (3). The reconstructed activity A was approximated by the activity in the VOI (C_{VOI}·V_{VOI}) scaled by the volume ratio V/V_{VOI}. The sphere diameter d was approximated as the effective VOI diameter (d_{eff} in Fig. 2, the diameter of a sphere with volume identical to the VOI volume) scaled by the factor v^{1/3}, where v is the ratio between the sphere volume and the VOI volume. In total, the concentration
where
v was determined by first performing a fit in Origin Pro 2015 (OriginLab Corporation, USA) of the function
to data for the five largest spheres in P_{∞} for each threshold method. This function was chosen for its asymptotic behavior, approaching V_{0} for small V and c·V for large V. The constraint 0 < c ≤ 1 was imposed. Then V(V_{VOI}) was determined from a lookuptable generated on basis of the fitted parameters, and
was found. When V_{VOI} < V_{0} the corresponding data point was excluded from this analysis.
Volume and background dependent calibration factors
For completeness, we also carried out analysis following the procedure proposed by Raskin et al. [17], in order to investigate if this procedure would be feasible for the QSPECT protocol within the range of spheretobackground ratios applied here. Only the unfiltered QSPECT reconstruction was considered as this was most analogous to the reconstruction used by Raskin et al. The calibration factor CF, i.e., the count rate for a VOI divided by the known activity in that VOI, was determined for each sphere as
where C_{true} is the known activity concentration in the sphere and S = 10.9 s^{−1} MBq^{−1} is the sensitivity determined from the QSPECT calibration procedure [20]. The socalled SpheretoBackground Voxel Ratio (SBVR) is equivalent to C_{dia}/C_{bkg} and was determined as such. CF versus SBVR and CF versus sphere volume was plotted and compared to the corresponding figures in Ref. [17].
Results
Maximum and peak values and accuracy
The maximum and the peak values relative to the mean concentration of the spheres are shown for all spheres and phantoms in Fig. 3, while the deviation from unity ratio between reconstructed and actual activity is shown in Fig. 4.
Contrast and recovery coefficient
The contrast and the recovery coefficients are shown for all spheres and phantoms in Fig. 5. The fitted values of α and β in Eq. (2) are (α; β) = (17.1 mm; 2.15), (22.2 mm; 2.21) and (20.6 mm; 1.63) for the QSPECT, the fQSPECT and the Bayesian reconstruction, respectively.
Mean concentration for spherical VOIs and thresholdbased VOIs
The mean concentrations in the spherical VOIs relative to the actual concentration are shown for all spheres in all phantoms in Fig. 6.
For the thresholdbased VOIs, the uncorrected mean to actual concentration is shown for all spheres in all phantoms in Figs. 7a, c, e and 8a, c, e and the corresponding volumes of the VOIs are shown in Figs. 7b, d, f and 8b, d, f.
The fit of Eq. (5) resulted in values of V_{0} in units of ml of (0.656; 1.157; 1.281; 1.773) for the QSPECT reconstruction, (1.966; 3.129; 2.444; 3.457) for the fQSPECT reconstruction and (3.492; 4.499; 4.229; 3.978) for the Bayesian reconstruction given in the order (50% max, 40% max, 60% peak, 50% peak)method. The values of V_{1} in units of ml are (48.9; 41.6; 55.9; 51.1), (60.6; 36.7; 84.9; 60.6) and (31.4; 9.5; 43.8; 3.5), respectively. The values of c are equal to the boundary of 1, except for the 40% max and 50% peak methods for the Bayesian reconstruction, where c is 0.776 and 0.584, respectively. Representative fits are shown in Figs. 7b, d, f and 8b, d, f.
The results for the mean values with correction for the partial volume effect using Eqs. (3)–(6) are shown in Figs. 9 and 10, respectively.
Volume and background dependent calibration factors
The calibration factors are shown as a function of SBVR and VOI volume in Fig. 11 for the QSPECT reconstruction.
Discussion
Phantom study
At the outset this study was planned with only the routinely applied QSPECT reconstruction without postfiltering [20, 23] in order to eventually apply the same reconstruction for tumor dosimetry as for kidney dosimetry. The 'valley' artifact observed in Fig. 1a, however, reduces the maximum and the peak values in the ø37 sphere to values below those of the ø28 sphere as shown in Fig. 3a. This carries on to lower the fixedpercentage thresholds and in turn influences values derived from the thresholdgenerated VOIs. The artifact can probably be ascribed to Gibbslike ringing artifacts in performing pointspread function correction as discussed by Kangasmaa et al. [21]. The artifact is not visible in Fig. 1d–f, where the transition from spheres to background is less sharp due to the presence of ^{177}Lu in the background volume. The smoothing by a Gaussian postfilter removes the artifact, and a more regular behavior is found in Fig. 1b and in Fig. 3c, d, where the mean and peak values for the ø37 sphere are larger than or equal to the values for the other spheres. With the Bayesian reconstruction, the voxel values are similar in voxels with similar Hounsfield Units in the CT scan. Indeed, in Fig. 1c, the voxel values across the spheres vary less than in the OSEM reconstructions and there is a sharp edge to the background.
As shown in Fig. 4, the accuracy is very good for the five largest spheres using the QSPECT reconstruction, which also means that despite the 'valley'artifact, the total counts are maintained within the ø37 sphere. For the fQSPECT reconstruction, the negative deviation is due to counts found outside the VOI_{dia+15} spheres as a consequence of the smoothing. For the Bayesian reconstruction the deviation is within 10% over the entire range of sphere sizes. A slightly better agreement may be obtainable by performing a full calibration for the Bayesian reconstruction rather than performing a scaling based on the unfiltered QSPECT reconstruction, however, this was not explored as the scaling approach was judged satisfactory for the present work.
The contrast shown in Fig. 5 is 100% for all reconstructions and all sphere sizes for P_{∞}, however, once background activity is introduced the values drop below the theoretical values in particular for the smallest spheres. Visually judged from the reconstructed images, the presence of a sphere, or tumor, cannot be recognized once the contrast is below about 0.2 (compare Figs. 1d–f and 5). The RCs in Fig. 5 show a regular behavior for P_{∞}, but for the other phantoms the presence of activity in the background volume causes the RCs of the smallest spheres to become larger than the RCs of some of the larger spheres. For the Bayesian reconstruction the picture is even inverted, such that the RCs for P_{2.7} and P_{5.0} are generally above those of P_{9.5} and P_{∞}.
One method for calculating tumor dose is based on the mean concentration in small spherical VOIs, for which data are shown in Fig. 6. The mean is within about 40% of the true value only for the largest sphere using the QSPECT reconstruction, but for the two largest spheres using the fQSPECT reconstruction and for the three largest using the Bayesian reconstruction. The variation of the obtained mean with ^{177}Lu content in the background is largest for the QSPECT reconstruction and smallest for the Bayesian reconstruction. The variation with chosen small VOI diameter can be observed from the series of three displaced data points for the same phantom and sphere diameter. Generally the mean decreases with increasing VOI diameter, as expected from the profiles in Fig. 1, where the voxel values vary across the sphere, being largest in the center, except in the presence of the 'valley'artifact. For the smaller spheres the mean values are generally quite low as compared to the true value, which can be explained by the low RCs and that the spherical VOIs probe a large fraction of the sphere volume.
For the method based on thresholdgenerated VOIs, the uncorrected data presented in Figs. 7a, c, e and 8a, c, e show that for the QSPECT reconstruction, values within 50% of the true value are obtained for the two largest spheres, while the mean is generally underestimated for the other two reconstructions except for the ø37 sphere. For the QSPECT reconstruction with a 40% of maximum or 50% of peak threshold, values within 40% of the true value are obtained for the two largest spheres for all phantoms except P_{2.7}. The mean values in VOIs generated by a percentage of the peak value were expected to be less susceptible to noise than those generated from the maximum in a single voxel, however, no appreciable difference in standard deviation of the mean is seen between the two. The volumes probed by the VOIs, as shown in Figs. 7b, d, f and 8b, d, f, vary significantly with the level of ^{177}Lu background activity, and in particular for the Bayesian reconstruction the volume may extend outside the sphere volume.
A model for partial volume correction was presented, and to this end the function in Eq. (5) was fitted to the volumes V_{VOI} vs V. The parameter c is not well determined, probably due to the lack of data points in the asymptotic limit of large volumes, and in most cases it is limited to the boundary of 1. Opportunely, the most important point is a good approximation at intermediate values of sphere volume, or diameter, where the slope of RC(d) is largest, which is also within the range where data points were obtained. Equation (5) was fitted to data points for P_{∞}, and as V_{VOI} was found to increase with background activity concentration, the scale factor v will be overestimated in the presence of background activity. This implies an overestimate of RC(d') and hence an underestimate of the correction 1/RC(d'). Additionally, the decrease found in RC(d) for increasing background activity for the OSEM reconstructions in Fig. 5 also leads to an underestimate of the correction, when RC(d) for P_{∞} is applied. One might argue, that by lowering the threshold one could obtain thresholdbased VOIs with volume equal or close to the sphere volumes, in order to enable partial volume correction directly on the basis of the VOI volume. This is true for P_{∞}, but in the presence of background activity or in the clinical case of an inhomogeneous background, the VOIs would in many cases grow unbounded by the sphere or tumor.
With correction for the partial volume effect, as presented in Figs. 9 and 10, the QSPECT reconstruction performs not very well with deviations between 50 and 100% even for the ø22 and ø28 spheres in P_{∞}. The Bayesian reconstruction on the other hand is very stable across all sphere sizes as compared to the other reconstructions. The values for the fQSPECT reconstruction with a threshold at 40% of the maximum or 50% of the peak value are within 40% of the true value for P_{∞} with a sphere diameter of 22 mm and larger. This is in line with the procedure applied by Ilan et al. [8], using a 42% threshold and a similar QSPECT procedure, with the same number of iterations and subsets in the OSEM reconstruction as here and a Hanning filter with cutoff at 0.85 cycles/cm. Once ^{177}Lu activity is introduced to the background volume, the corrected mean drops significantly for the ø22 sphere (P_{9.5}) or the thresholdbased VOI even extends beyond limits (P_{5.0} and P_{2.7}), while for the ø28 and ø37 spheres the mean remains within about 40% of the true value.
In Fig. 11a a linear increase of CF as a function of SBVR was anticipated on the basis of Fig. 4 in the paper by Raskin et al. [17], but this is clearly not the case. Apparently this linear behavior breaks down for the lower spheretobackground ratio applied here (actual 2.7–9.5; SBVR 1.3–6.2 for the three largest spheres) as compared to Ref. [17] (actual 6–17; SBVR 2.4–14.6)—at least for the applied acquisition and reconstruction parameters. For the two smallest spheres, not analyzed in Ref. [17], a marked decrease of CF with SBVR can even be observed. The behavior of CF with respect to volume as shown in Fig. 11b is similar to the measured data points in Fig. 5 of Ref. [17], however, the CF is found to increase with decreasing SBVR (due to spillin), while the opposite behavior is seen in Ref. [17], possibly due to the larger number of OSEM iteration updates (100 vs. 32 in the present work). We further note that the CF is strongly dependent on volume for the considered sphere volumes, which would make a correction rather sensitive to the determined volume of a given lesion. With the significant differences found in comparison to Ref. [17], further investigations would be needed to apply the method of Raskin et al. for correction of volume and backgrounddependence.
Translation to tumor dosimetry
For tumor dosimetry, we assume that the mean tumor dose is calculated from the mean ^{177}Lu activity concentration in a tumor in sequential posttreatment scans, and that the betaradiation is completely absorbed within the tumor.
When the results of the phantom study are considered in the context of tumor dosimetry, some options and limitations become clear. The data obtained for P_{∞} in Fig. 4 indicate, that the ^{177}Lu activity in a wellisolated tumor even down to 13 mm diameter can be determined to within about 10% using a large VOI around the tumor on the QSPECT reconstruction. In clinical cases, typical quantification accuracy of ^{177}Lu activity is about 10–20% [20, 23, 24], and provided the tumor volume is determined with a similar accuracy using other imaging methods, e.g. PET, CT or MR, a similarly accurate measure of ^{177}Lu concentration can be obtained.
The methods based on a small VOI or a thresholdgenerated VOI can be applied not only for isolated tumors, but also if nearby activity is present, e.g. in other tumors or as physiological uptake in liver, spleen or kidneys. For the small VOI method and the threshold method with partial volume correction, the best agreement is found for the two or three largest spheres for the fQSPECT and the Bayesian reconstruction, respectively, with the threshold set to 40% of the maximum or 50% of the peak value. We also observed that for the QSPECT reconstruction a threshold set to 40% of the maximum or 50% of the peak value leads to equally good agreement without partial volume correction. This is an empirical observation, and the method can be seen as an alternative to the small VOI method for selecting a representative volume. For practical reasons it could be an attractive option, as it makes use of the QSPECT reconstruction, which already is applied for kidney dosimetry.
By the nature of the Bayesian reconstruction, it critically depends on the alignment between SPECT and CT, and hence the relatively good performance in this phantom study cannot necessarily be translated to patient studies, e.g. an 'erratic fragmented appearance' of lesions was reported by Grootjans et al. [25], and the Bayesian reconstruction should only be used with caution. Hence, the small VOI or the thresholdgenerated VOI method are judged more robust and straightforward to apply using an OSEM reconstruction.
As a practical limit we suggest for both the small VOI method and the thresholdbased methods a tumor diameter > 30 mm. This is slightly larger than the ø28 sphere and thus a bit conservative to allow for some deviations from the idealized case of a phantom study. Tumors down to 22 mm diameter can also be included for analysis, if they are well isolated, using either the smallVOI method with a ø10 VOI or the thresholdbased methods. Similarly for the large VOI method, we slightly conservatively require a tumor diameter larger than 15 mm. Tumor diameters are often reported as the longest diameter in the plane of measurement [26], while partial volume effects are expected to be most significant in the plane of the shortest axis of nonspherical volumes. Hence if the aspect ratio differs significantly from unity, e.g. as for very elongated tumors, the short axis should also fulfill the relevant limit. It is wellknown from phantom studies of kidneys that the recovery coefficient decrease with increasing aspect ratio, or surfacetovolume ratio, as well as with the complexity of the geometry [27, 28], and similar results would be expected from a similar study of nonspherical tumor geometries.
For all ø28 and ø37 spheres in all phantoms the measured contrast is > 0.2. This is, however, also at the limit where a tumor can be recognized, and as a practical limit for the small VOI method we propose a contrast > 0.3, or equivalently a tumortobackground ratio larger than 2. Some variation of the mean concentration with small VOI diameter is found, particularly when no ^{177}Lu background activity is present. A ø15 or ø20 diameter VOI may be preferable, as the concentration tends to be overestimated with a ø10 VOI, but a general optimal choice cannot be made as this depends on tumor diameter. A 20 mm diameter sphere was applied by Sandström and colleagues in kidney and spleen dosimetry [29], and adapted for tumor dosimetry by Del Prete et al. [11, 30]. The thresholdbased methods (40% of maximum or 50% of peak value) could only be applied to spheres for which the measured contrast was > 0.5 or equivalently the tumortobackground ratio was larger than 3. The tumortobackground ratio is likely to vary in the days following treatment, and the contrast criteria must be fulfilled for all posttreatment scans used for tumor dosimetry. Normally the treatment drug is retained longer in tumors than in organs, which means of course that if the tumortobackground criterion is fulfilled at the first posttreatment scan, it is most likely also fulfilled at later scans.
With these criteria for selecting tumors for analysis by one of the three methods, it should be possible to determine the mean ^{177}Lu concentration at the level of 40% accuracy, or better with the large VOI method. Other significant error sources can be the volume determination (large VOI method), the partial volume correction (thresholdgenerated VOI method) or errors in fitting sequential data points to an exponential decay [20]. In clinical practice, quantification errors larger than the 40% found here cannot be excluded, and as further error sources also contribute to the total error of the tumor dose, a realistic boundary on the possible error is not below 50%. If the estimated tumor dose is within 50% of the true value, then the true value is within + 100%/− 33% of the estimated dose.
This seems perhaps unsatisfactory, but since tumor dosimetry is more difficult than kidney dosimetry, with its smaller volumes that are undiscernible on noncontrast CT, errors up to 50% are not surprising [10, 31]. For a symmetric triangular error distribution, a 50% maximal error corresponds to a standard deviation of 20% [20, 32], which can be compared to reported standard deviations in kidney dosimetry of 10–15% [20, 31].
This level of uncertainty in tumor dosimetry is lower than observed interpatient or intertumor variability or the intrapatient variability through the course of a treatment series [11, 14, 33], and further the thresholdbased method with partial volume correction has already been used to demonstrate tumordose response for pancreatic neuroendocrine tumors [8]. This holds promises that any of the methods with their associated restrictions are useful for e.g. dose response studies in PRRT and RLT using ^{177}Lu.
These conclusions are based on a phantom study with spheres up to 37 mm diameter, but are also expected to hold for larger tumors, where partial volume effects are less significant [7]. The QSPECT reconstruction method is applied at least at three sites, including our own [11, 20, 34]. Overall similar conclusions are expectable for other hardware, acquisition or reconstruction parameters [35], but the limits on tumor diameter and tumortobackground ratio as well as the choice of threshold or small VOI size may differ. A consistency check of the methods can be made for tumors, when more than one of the methods are applicable. Sandström et al. [29] demonstrated a good correspondence between the small VOI and the thresholdbased method in dosimetry of the kidneys and the spleen.
Conclusion
Determination of the mean ^{177}Lu concentration in a tumor, as represented here by a sphere, is challenging and depends on factors such as tumor volume, reconstruction parameters, background activity and possibly artifacts. Using either

the large VOI method on the QSPECT reconstruction, with tumor volume from other modalities, for solitary tumors larger than 15 mm diameter,

the small VOI method on the fQSPECT reconstruction for tumors larger than 30 mm diameter and with tumortobackground ratio > 2, or

the thresholdbased method (40% of maximum or 50% of peak value) without partialvolume correction on the QSPECT reconstruction, or with partial volume correction on the fQSPECT reconstruction, for tumors larger than 30 mm diameter and tumortobackground ratio > 3,
a determination of mean ^{177}Lu concentration in tumors with background activity can be made, which is expected to be useful in e.g., dose response studies and tumor to normal organ dose comparisons. None of the methods require accurate tumor delineation, and the determined concentration does not depend directly on tumor volume or tumortobackground ratio—only the abovementioned limits should be fulfilled. The methods are wellknown and straightforward to implement using standard tools in commercially available software.
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the author on reasonable request.
Abbreviations
 DTPA:

Diethylenetriaminepentaacetic acid
 IEC:

International Electrotechnical Commission
 NEMA:

National Electrical Manufacturers Association
 OSEM:

Orderedsubsetexpectation–maximization
 PRRT:

Peptide receptor radionuclide therapy
 PSMA:

Prostatespecific membrane antigen
 QSPECT:

Quantitative SPECT
 RC:

Recovery coefficient
 RLT:

Radioligand therapy
 VOI:

VolumeofInterest
References
Bodei L, MuellerBrand J, Baum RP, Pavel ME, Horsch D, O’Dorisio MS, et al. The joint IAEA, EANM, and SNMMI practical guidance on peptide receptor radionuclide therapy (PRRNT) in neuroendocrine tumours. Eur J Nucl Med Mol Imaging. 2013;40(5):800–16.
Ambrosini V, Kunikowska J, Baudin E, Bodei L, Bouvier C, Capdevila J, et al. Consensus on molecular imaging and theranostics in neuroendocrine neoplasms. Eur J Cancer. 2021;146:56–73.
Kratochwil C, Fendler WP, Eiber M, Baum R, Bozkurt MF, Czernin J, et al. EANM procedure guidelines for radionuclide therapy with ^{177}Lulabelled PSMAligands (^{177}LuPSMARLT). Eur J Nucl Med Mol Imaging. 2019;46(12):2536–44.
LawhnHeath C, Hope TA, Martinez J, Fung EK, Shin J, Seo Y, et al. Dosimetry in radionuclide therapy: the clinical role of measuring radiation dose. Lancet Oncol. 2022;23(2):e75–87.
Cremonesi M, Ferrari ME, Bodei L, Chiesa C, Sarnelli A, Garibaldi C, et al. Correlation of dose with toxicity and tumour response to ^{90}Y and ^{177}LuPRRT provides the basis for optimization through individualized treatment planning. Eur J Nucl Med Mol Imaging. 2018;45(13):2426–41.
Jackson P, Hofman M, McIntosh L, Buteau JP, Ravi KA. Radiation dosimetry in (177)LuPSMA617 therapy. Semin Nucl Med. 2022;52(2):243–54.
Ljungberg M, SjogreenGleisner K. The accuracy of absorbed dose estimates in tumours determined by quantitative SPECT: a Monte Carlo study. Acta Oncol. 2011;50(6):981–9.
Ilan E, Sandstrom M, Wassberg C, Sundin A, GarskeRoman U, Eriksson B, et al. Dose response of pancreatic neuroendocrine tumors treated with peptide receptor radionuclide therapy using ^{177}LuDOTATATE. J Nucl Med. 2015;56(2):177–82.
Jahn U, Ilan E, Sandstrom M, GarskeRoman U, Lubberink M, Sundin A. ^{177}LuDOTATATE peptide receptor radionuclide therapy: dose response in small intestinal neuroendocrine tumors. Neuroendocrinology. 2020;110(7–8):662–70.
Finocchiaro D, Gear JI, Fioroni F, Flux GD, Murray I, Castellani G, et al. Uncertainty analysis of tumour absorbed dose calculations in molecular radiotherapy. EJNMMI Phys. 2020;7(1):63.
Del Prete M, Buteau FA, Arsenault F, Saighi N, Bouchard LO, Beaulieu A, et al. Personalized ^{177}Luoctreotate peptide receptor radionuclide therapy of neuroendocrine tumours: initial results from the PPRRT trial. Eur J Nucl Med Mol Imaging. 2019;46(3):728–42.
Okamoto S, Thieme A, Allmann J, D’Alessandria C, Maurer T, Retz M, et al. Radiation dosimetry for ^{177}LuPSMA I&T in metastatic castrationresistant prostate cancer: absorbed dose in normal organs and tumor lesions. J Nucl Med. 2017;58(3):445–50.
Peters SMB, Prive BM, de Bakker M, de Lange F, Jentzen W, Eek A, et al. Intratherapeutic dosimetry of [^{177}Lu]LuPSMA617 in lowvolume hormonesensitive metastatic prostate cancer patients and correlation with treatment outcome. Eur J Nucl Med Mol Imaging. 2022;49(2):460–9.
Ezziddin S, Reichmann K, YongHing C, Damm M, Risse J, Ahmadzadehfar H, et al. Early prediction of tumour response to PRRT. The sequential change of tumourabsorbed doses during treatment with 177Luoctreotate. Nuklearmedizin. 2013;52(5):170–7.
Goetz TI, Lang EW, Prante O, Maier A, Cordes M, Kuwert T, et al. Threedimensional Monte Carlobased voxelwise tumor dosimetry in patients with neuroendocrine tumors who underwent ^{177}LuDOTATOC therapy. Ann Nucl Med. 2020;34(4):244–53.
Roth D, Gustafsson J, Sundlov A, Sjogreen GK. A method for tumor dosimetry based on hybrid planarSPECT/CT images and semiautomatic segmentation. Med Phys. 2018;45(11):5004–18.
Raskin S, Gamliel D, Abookasis D, BenHaim S, Chicheportiche A. Towards accurate ^{177}Lu SPECT activity quantification and standardization using lesiontobackground voxel ratio. EJNMMI Phys. 2023;10(1):5.
National Electrical Manufacturers Association (NEMA). Performance Measurements of Positron Emission Tomographs. NEMA Standards Publication NU2–2001. National Electrical Manufacturers Association 2001.
Park MA, Mahmood A, Zimmerman RE, LimpaAmara N, Makrigiorgos GM, Moore SC. Adsorption of metallic radionuclides on plastic phantom walls. Med Phys. 2008;35(4):1606–10.
Staanum PF, Frellsen AF, Olesen ML, Iversen P, Arveschoug AK. Practical kidney dosimetry in peptide receptor radionuclide therapy using [^{177}Lu]LuDOTATOC and [^{177}Lu]LuDOTATATE with focus on uncertainty estimates. EJNMMI Phys. 2021;8(1):78.
Kangasmaa T, Sohlberg A, Kuikka JT. Reduction of collimator correction artefacts with Bayesian reconstruction in SPECT. Int J Mol Imaging. 2011;2011: 630813.
Gear JI, Cox MG, Gustafsson J, Sjögreen Gleisner K, Murray I, Glatting G, et al. EANM practical guidance on uncertainty analysis for molecular radiotherapy absorbed dose calculations. Eur J Nucl Med Mol Imaging. 2018;45:2456–74.
Beauregard JM, Hofman MS, Pereira JM, Eu P, Hicks RJ. Quantitative (177)Lu SPECT (QSPECT) imaging using a commercially available SPECT/CT system. Cancer Imaging. 2011;11:56–66.
Kennedy J, Chicheportiche A, Keidar Z. Quantitative SPECT/CT for dosimetry of peptide receptor radionuclide therapy. Semin Nucl Med. 2022;52(2):229–42.
Grootjans W, Meeuwis AP, Slump CH, de GeusOei LF, Gotthardt M, Visser EP. Performance of 3DOSEM and MAP algorithms for reconstructing low count SPECT acquisitions. Z Med Phys. 2016;26(4):311–22.
Eisenhauer EA, Therasse P, Bogaerts J, Schwartz LH, Sargent D, Ford R, et al. New response evaluation criteria in solid tumours: revised RECIST guideline (version 1.1). Eur J Cancer. 2009;45(2):228–47.
TranGia J, Lassmann M. Optimizing image quantification for (177)Lu SPECT/CT based on a 3D printed 2compartment kidney phantom. J Nucl Med. 2018;59(4):616–24.
Grings A, Jobic C, Kuwert T, Ritt P. The magnitude of the partial volume effect in SPECT imaging of the kidneys: a phantom study. EJNMMI Phys. 2022;9(1):18.
Sandström M, Ilan E, Karlberg A, Johansson S, Freedman N, GarskeRoman U. Method dependence, observer variability and kidney volumes in radiation dosimetry of (177)LuDOTATATE therapy in patients with neuroendocrine tumours. EJNMMI Phys. 2015;2(1):24.
Del Prete M, Buteau FA, Beauregard JM. Personalized (177)Luoctreotate peptide receptor radionuclide therapy of neuroendocrine tumours: a simulation study. Eur J Nucl Med Mol Imaging. 2017;44(9):1490–500.
Gear JI, Cox MG, Gustafsson J, Gleisner KS, Murray I, Glatting G, et al. EANM practical guidance on uncertainty analysis for molecular radiotherapy absorbed dose calculations. Eur J Nucl Med Mol Imaging. 2018;45(13):2456–74.
Joint Committee for Guides in Metrology. JCGM 100. Evaluation of measurement data—guide to the expression of uncertainty in measurement. Sèvres: BIPM; 2008.
Violet J, Jackson P, Ferdinandus J, Sandhu S, Akhurst T, Iravani A, et al. Dosimetry of (177)LuPSMA617 in metastatic castrationresistant prostate cancer: correlations between pretherapeutic imaging and wholebody tumor dosimetry with treatment outcomes. J Nucl Med. 2019;60(4):517–23.
Jackson PA, Hofman MS, Hicks RJ, Scalzo M, Violet J. Radiation dosimetry in (177)LuPSMA617 therapy using a single posttreatment SPECT/CT scan: a novel methodology to generate time and tissuespecific dose factors. J Nucl Med. 2020;61(7):1030–6.
Peters SMB, Meyer Viol SL, van der Werf NR, de Jong N, van Velden FHP, Meeuwis A, et al. Variability in lutetium177 SPECT quantification between different stateoftheart SPECT/CT systems. EJNMMI Phys. 2020;7(1):9.
Acknowledgements
Not applicable.
Funding
None.
Author information
Authors and Affiliations
Contributions
PFS designed and performed the study and wrote the manuscript.
Corresponding author
Ethics declarations
Ethics approval and consent to participate
Not applicable.
Consent for publication
Not applicable.
Competing interests
The author declares that he has no competing interests.
Additional information
Publisher's Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made. The images or other third party material in this article are included in the article's Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article's Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this licence, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Staanum, P.F. Tumor dosimetry using ^{177}Lu: influence of background activity, measurement method and reconstruction algorithm. EJNMMI Phys 10, 39 (2023). https://doi.org/10.1186/s40658023005618
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/s40658023005618
Keywords
 Radionuclide therapy
 Tumor dosimetry
 Lu177
 Phantom