- Original Research
- Open Access
Fully automated calculation of image-derived input function in simultaneous PET/MRI in a sheep model
- Thies H. Jochimsen†1Email author,
- Vilia Zeisig1,
- Jessica Schulz2,
- Peter Werner1,
- Marianne Patt1,
- Jörg Patt1,
- Antje Y. Dreyer3, 4,
- Johannes Boltze3, 4, 5,
- Henryk Barthel1,
- Osama Sabri1 and
- Bernhard Sattler1
© Jochimsen et al. 2016
- Received: 1 October 2015
- Accepted: 29 January 2016
- Published: 13 February 2016
Obtaining the arterial input function (AIF) from image data in dynamic positron emission tomography (PET) examinations is a non-invasive alternative to arterial blood sampling. In simultaneous PET/magnetic resonance imaging (PET/MRI), high-resolution MRI angiographies can be used to define major arteries for correction of partial-volume effects (PVE) and point spread function (PSF) response in the PET data. The present study describes a fully automated method to obtain the image-derived input function (IDIF) in PET/MRI. Results are compared to those obtained by arterial blood sampling.
To segment the trunk of the major arteries in the neck, a high-resolution time-of-flight MRI angiography was postprocessed by a vessel-enhancement filter based on the inertia tensor. Together with the measured PSF of the PET subsystem, the arterial mask was used for geometrical deconvolution, yielding the time-resolved activity concentration averaged over a major artery. The method was compared to manual arterial blood sampling at the hind leg of 21 sheep (animal stroke model) during measurement of blood flow with O15-water. Absolute quantification of activity concentration was compared after bolus passage during steady state, i.e., between 2.5- and 5-min post injection. Cerebral blood flow (CBF) values from blood sampling and IDIF were also compared.
The cross-calibration factor obtained by comparing activity concentrations in blood samples and IDIF during steady state is 0.98 ± 0.10. In all examinations, the IDIF provided a much earlier and sharper bolus peak than in the time course of activity concentration obtained by arterial blood sampling. CBF using the IDIF was 22 % higher than CBF obtained by using the AIF yielded by blood sampling.
The small deviation between arterial blood sampling and IDIF during steady state indicates that correction of PVE and PSF is possible with the method presented. The differences in bolus dynamics and, hence, CBF values can be explained by the different sampling locations (hind leg vs. major neck arteries) with differences in delay/dispersion. It will be the topic of further work to test the method on humans with the perspective of replacing invasive blood sampling by an IDIF using simultaneous PET/MRI.
- Arterial input function
- Image-derived input function
To quantify metabolic or physiological function by means of of radiotracers in positron emission tomography (PET), kinetic models are applied to calculate the rate constants between different compartments, e.g., between blood and tissue. For these models, the measurement of the arterial input function (AIF) is mandatory. The AIF is the time-activity concentration curve of an injected radiopharmaceutical in arterial blood plasma that is delivered to the target tissue. This activity curve is normally determined by taking arterial blood samples from the patient at different time points after injection.
As a potential alternative, the AIF could also be obtained from the PET data itself, yielding an image-derived input function (IDIF). Different approaches have been taken to calculate the IDIF. Firstly, the IDIF can be derived from a blood pool (major vessels, ventricles of the heart) visible in the low-resolution PET data using mathematical modeling and/or a few blood samples for calibration [1–11]. Secondly, data from a co-registered high-resolution imaging technique can be used to segment the arterial blood pool. These can be magnetic resonance imaging (MRI) [12–15] as well as computed tomography (CT) images .
To obtain an accurate estimation of IDIF, the so-called spill-in and spill-out, partial-volume effects (PVE) have to be corrected for. For this correction, the exact knowledge about the point spread function (PSF) response of the PET imaging system and the geometry (diameter, orientation) of the artery is necessary. However, the co-registration of structures in sequentially acquired sessions in different imaging modalities may be difficult because, e.g., the carotid artery changes its exact position with the head position and global registration algorithms usually are based on other structures. Using hybrid PET/MRI scanners where the images can be acquired simultaneously allows for a precise segmentation of the artery .
To be useful in research and clinical practice, manual operation has to be minimized while generating the IDIF. Ideally, the IDIF is generated fully automated from the simultaneous PET/MRI data. This work describes such a procedure applied to a study with an animal model (sheep). It compares the generated IDIF with an AIF obtained by blood sampling to quantify cerebral blood flow (CBF) by 15O-H2O-PET imaging. Moreover, CBF values obtained by both methods are compared.
Animal model and imaging procedure
The animal experiments were conducted in accordance with the recommendations of the European Convention for the Protection of Vertebrate Animals used for Experimentation and the current ARRIVE guidelines. The animal experiments were approved by the local animal welfare authorities (Directorate Leipzig, Germany). The examinations were part of a preclinical stroke study, investigating therapeutic effects of stem cell application. For that purpose, adult merino sheep (n=21; 62 ± 8-kg body weight) underwent permanent proximal middle cerebral artery occlusion as described previously [18, 19]. Examinations were performed on a clinical PET/MRI scanner (Biograph mMR, Siemens Healthcare, Erlangen, Germany) after induction of ischemia (from 4 h to 35 days after stroke). MR signal was received by a flexible body surface coil wrapped over the animal head in addition to the spine coil which is integrated in the patient table. The surface coil is not accounted for in the attenuation correction. Imaging included structural (e.g., T1-weighted, diffusion-tensor imaging), diagnostic (e.g., fluid-attenuated inversion recovery, turbo spin echo), and functional (dynamic susceptibility contrast-based perfusion imaging, arterial spin labeling) MRI sequences.
A time-of-flight MRI angiography (TOF-MRA) sequence was used to identify the trunk of the major neck arteries. Parameters of the sequence were as follows: 0.5 × 0.5 × 0.7 mm 3 voxel size, 4 slabs, 40 slices per slab, TE = 3.6 ms, TR = 21 ms, 21° flip angle, GRAPPA = 2, saturation pulse for venous signal, and 70 % tilted optimized nonsaturating excitation pulse.
To determine CBF, sheep were subjected to a 5-min 15O-H2O-PET scan starting with a bolus injection of 1064 ± 238 MBq (mean and standard deviation over subjects) into the jugular vein. Blood samples were withdrawn manually from a femoral artery (left/right side depending on accessibility of arterial cannulation) during the PET scan. For the first 2 min after tracer injection, blood sampling was performed dynamically (approximately every 3 s), followed by predefined time points: 150, 180, 210, 240, and 300 s. Recording the sampling procedure with a video camera allowed for temporal synchronization to the start of PET scan as well as a retrospective definition of the exact sampling time points. The blood activity concentration was measured using a gamma counter (WIZARD2, PerkinElmer LAS GmbH, Rodgau, Germany) cross-calibrated to the PET/MRI scanner. After an upgrade of the blood sampling procedure, the simultaneous TOF-MRA/PET measurements (see below) were performed by an automated blood sampler (Twilite, Swisstrace GmbH, Menzingen, Switzerland) also cross-calibrated to the PET/MRI.
PET data was reconstructed into a 128 × 128 matrix (voxel size: 1.40 × 1.40 × 2.03 mm3) using the built-in 3D ordered subset expectation maximization (OSEM) algorithm with 8 iterations, 21 subsets, and a 3-mm Gaussian filter. Scatter and Dixon-based attenuation correction were applied. The following consecutive time frames were used for the dynamic analysis: 20 × 3, 12 × 5, 12 × 10, and 2 × 30 s.
Segmentation of major arteries
To segment only the major arteries, clusters of adjacent voxels are generated (step C in Fig. 1). The two largest clusters, which correspond to the straight parts of the bilateral carotid arteries, are selected automatically (step D in Fig. 1). Finally, to avoid bias from the extended peripheral venous catheter which was used for injection of 15O-water (Fig. 5), only the artery on the contralateral side was used, i.e., furthest from the catheter. This artery was selected automatically based on the maximum distance of its center of gravity (COG) (arterial mask interpreted as discrete masses) from the PET COG (maximum PET bolus activity concentration interpreted as discrete masses).
Although a simultaneous PET-MRI system was used which provides inherent co-registration, the TOF-MRA data for segmentation of arteries was acquired approximately 20 to 30 min prior to the PET acquisition. This offset was dictated by other constrains of the overall study.
Although animals were anesthetized, we observed a spatial misalignment between TOF-MRA and PET, probably caused by slow sinking/rolling of the animal during this period of time. This misalignment was corrected for by the following procedure (step E in Fig. 1). Because the principal direction of the arteries is in axial direction, it was assumed that the major impact on results is caused by motion perpendicular to the axial direction. Thus, motion correction was performed slice-by-slice in a two-dimensional fashion. In each transaxial slice, the COG of the segmented arterial voxels was compared with the position of the maximum of PET signal. This two-dimensional shift was fitted as a function of slice position to a second-order polynomial to obtain a smooth slice-to-slice transition. Finally, the fitted shift was used in a slice-by-slice spline interpolation to align the artery segmentation (binary mask) to the PET images (step F in Fig. 1). As described previously , this procedure yielded a partial-volume coefficient (values between 0 and 1) assigned to each PET voxel which reflects the fractional overlap with the TOF-MRA mask.
To test whether the alignment procedure would be necessary in a truly simultaneous TOF-MRA and PET examination, 4 sheep were investigated with the TOF-MRA sequence acquired during the PET scan.
Partial-volume effect and point-spread correction
Having identified and aligned the arteries with high resolution, the next step is to compensate for spill-over effects in the low-resolution PET data. For that, the PSF has to be known. To obtain the PSF of the PET imaging modality (detector hardware and reconstruction algorithm), the following calibration experiment was performed once (G in Fig. 1): a 1-ml syringe with a known diameter of 4.7 mm (cylinder length: 57 mm) was filled with 95-MBq 18F-fludeoxyglucose. The syringe was placed axially in the scanner in approximately the same off-center position (50 mm) where the TOF-MRA volume would be in the actual 15O-H2O-PET/MRI examination. PET data was acquired and reconstructed in the same way as in the actual examinations. For 16 axial slices, which were perpendicular to the axis of the syringe and included a cross-sectional image in the central part of the syringe (i.e., without edge effects), the following procedure was applied to estimate the full-width at half-maximum (FWHM) of the PSF. A model function was fitted to each slice using a downhill simplex algorithm. The function was composed of a disk shape (syringe cross-section, i.e., 1 inside the syringe and 0 outside) convolved with a Gaussian (representing the PSF) and allowed a variable position inside the imaging plane. Four independent parameters of the function were used in the fitting procedure: the two-dimensional position of the disk, the amplitude, and the FWHM of the Gaussian. Averaging over all slices yielded an average FWHM.
The data from the PSF experiment was also used to estimate the bias in activity concentration as a function of different frame durations which might be introduced by the OSEM reconstruction . For this estimation, a mask containing only the syringe was generated by the same histogram-based algorithm as used for segmentation of arteries (see above). For each time frame, activity was averaged over all voxels contained in the mask. Finally, it was normalized to the mean over time.
Convolving the partial-volume map (as obtained by step F in Fig. 1) with the PSF yielded a partial-volume coefficient per voxel which also takes PSF effects (spill-in and spill-out) into account (step H in Fig. 1). Using this coefficient, the activity concentration of all voxels in a predefined region (all voxels not further away than two FWHM from the next arterial voxel) was modeled by a linear combination of intra- and extraarterial concentration. This geometric transformation matrix , which is a massively overdetermined set of linear equations, was inverted and applied to each time frame (step I in Fig. 1), yielding the IDIF and the extraarterial activity concentration.
Comparison of IDIF and AIF
In addition to the cross-calibration factor, the ratio of intra- vs. extraarterial (tissue surrounding vessel) activity concentration, as obtained by the geometrical deconvolution described above, was evaluated in steady state. Furthermore, the area under curve (AUC) of IDIF and AIF was compared for the whole scan (0–300 s).
CBF maps were calculated using either AIF or IDIF for kinetic modeling with PMOD software (PMOD Technologies Ltd, Zurich, Switzerland) applying the weighted integration method according to Alpert et al.  using the whole 5-min scan. In case of AIFs, where the blood tracer activity concentration is recorded with an external device, the measured activity concentration is distorted relative to the activity signal arriving in the brain by two effects: a tracer time delay and a broadening of the peak of the AIF (bolus dispersion). For exact CBF measurements, the delay and dispersion of the blood measurements must, therefore, be corrected for. In our study, the PMOD-implemented method by Meyer  was used. CBF values were averaged over a manually drawn region covering the healthy hemisphere (i.e., not directly affected by stroke) for each subject. In the group with simultaneous TOF-MRA and PET, one sheep had to be excluded from the CBF analysis because of missing arterial blood data during bolus passage.
The transaxial spatial shift in the alignment procedure, averaged over all slices, was 3.18 ± 1.69 mm (mean and standard deviation over subjects) with a maximum of 8.94 mm for the consecutive TOF-MRA/PET. For the simultaneous TOF-MRA/PET, it was 2.02 ± 0.57 mm with a maximum of 2.66 mm.
The segmented arteries of all sheep from the consecutive TOF-MRA and PET examinations are shown in Figs. 4 and 5. Visually, good segmentation of arteries can be observed. The algorithm reliably detects the artery on the opposite side of the catheter to reduce bias from the catheter.
Parameters of IDIF and AIF for consecutive and simultaneous TOF-MRA/PET acquisition
AUC IDIF/AUC AIF
AUC IDIF/AUC AIF
According to Table 1, the intraarterial activity concentration in steady state is approximately three times higher than in the extraarterial space. It is worth noting that this factor is significantly increased by the spatial alignment for the consecutive TOF-MRA/PET data, while it remains almost the same for the simultaneous TOF-MRA/PET. Taking the consecutive and simultaneous TOF-MRA/PET together, the AUC of the AIF is approximately 5 % higher than the AUC of the IDIF for the spatially aligned data (labeled ‘Shifted’ in Table 1).
CBF values obtained by IDIF and AIF for consecutive and simultaneous TOF-MRA/PET acquisition
Comparing the aligned vs. the unaligned data, a major improvement can be seen for the consecutive TOF-MRA/PET data by aligning the TOF-MRA data, i.e., the standard deviation of IDIF-based CBF is reduced by a factor of 5 when the TOF-MRA data is shifted to match spatial position of the arteries in PET. This improvement could not be seen in the simultaneous TOF-MRA/PET. Also, CBF values of simultaneous TOF-MRA/PET are more similar than for the consecutive TOF-MRA/PET. However, the extremely small standard deviation of the unshifted IDIF-based CBF of the simultaneous TOF-MRA/PET in Table 2 is most likely only a coincidence and not representative due to the small sample size (n=3). Hence, it is difficult to say whether a significant difference (e.g., as estimated by a t test) remains between simultaneous TOF-MRA/PET and motion-corrected consecutive TOF-MRA/PET.
Our results show that the combination of TOF-MRA, vessel-enhancement filter, and automated mask selection allows a robust segmentation of arteries. Another approach to enhance vessel-like structures based on the second-order directional derivatives (Hessian matrix)  might also be feasible instead of evaluating the inertia tensor. Interactive selection of a 3D seed point, as for instance in , is not required for the approach presented here. Thus, the present method allows a fully automated calculation of IDIF shortly after the data was acquired, allowing, for instance, rapid calculation of CBF maps. This is of importance for examinations where timely diagnosis is mandatory, e.g., in stroke.
There is good agreement in activity concentrations of blood samples and IDIF after bolus passage during steady state. Conversely, CBF values are significantly higher when based on IDIF. The most plausible explanation for this deviation are the different sampling locations (hind leg vs. major neck arteries) with differences in delay/dispersion (including their correction by PMOD in case of AIF) leading to different bolus dynamics.
One may argue that the good agreement in steady state is simply a consequence of a homogenous distribution of the tracer in the body far after bolus passage due to perfusion and extravasation of 15O-H2O into the extrarterial space, and that the differences in CBF (approximately 20 %) arise from an insufficient quantification of tracer concentration during bolus passage. However, as tissue CBF is inversely proportional to the arterial AUC, this effect may only account for approximately 5 % of the difference. Moreover, a three-times higher activity concentration in the intraarterial space compared to extraarterial space in steady state (cf. Table 1) indicates that the tracer is not homogenously distributed in steady state after bolus passage.
The measured activity concentration is relatively independent of the frame duration (cf. Fig. 8). Only the first frame shows a considerable bias. However, as the IDIF is negligibly small in the first frame, the impact on CBF analysis will be small.
Another source of quantification error could be an insufficient attenuation correction as the Dixon-based method does not account for bones. Moreover, the flexible surface coil is not accounted for in the attenuation correction. However, the good agreement in activity concentrations of blood samples and IDIF indicates that these sources of error have a minor impact on quantification. This might be because there are fewer bones in the neck compared to the brain/head. Also, the surface coil was distant from the neck.
The comparison of consecutive (i.e., with a delay of 20 to 30 min) with the simultaneous TOF-MRA/PET suggests that retrospective spatial alignment is a mandatory step in case that angiography and PET cannot be acquired truly simultaneously, e.g., when dictated by other constrains of the overall study, as in this work. Our data suggest that spatial registration is not necessary in simultaneous TOF-MRA/PET. However, a residual difference in CBF values before and after spatial alignment can be observed even in the simultaneous TOF-MRA/PET data. This difference can be explained by other effects. For instance, field distortions can cause spatial displacements in MRI. Also, the bolus passage at the beginning of the 5-min scan is the crucial time for the CBF analysis, whereas TOF-MRA is acquired slab-by-slab during the whole scan. Thus, subject motion during the scan can also cause differences in CBF.
An additional problem that may arise if the input function is calculated only from image data is the fact that the initial rapid rise and decay of the input function curve cannot be sampled very precisely from the PET data due to its low temporal resolution. On the other hand, by using a PET/MRI system with a relatively large axial PET FOV (258 mm) in this study, a large arterial volume is available within the FOV for the calculation of the IDIF. Therefore, more radioactive decays are present in the arterial volume under observation, and the temporal resolution of the IDIF can possibly be increased while maintaining a reasonable number of counts per sampling interval in the IDIF.
Unlike humans, sheep have an arterial structure (rete mirabile) consisting of a complex net of arteries close to the major neck arteries. This structure is visible in the early PET images, for instance, in S24 in Fig. 5 (indicated by a green arrow). In contrast, this structure is not visible in the TOF-MRA data. This is most likely because the TOF-MRA method is only sensitive to blood flowing into the imaging slab with a relatively high velocity. Therefore, the blood in this structure may bias quantification by uncorrected spill-over effects in PET. However, the relatively good agreement of blood samples and IDIF in steady state suggests that this structure is not a major source of error in the present study.
It should be straightforward to extend the present method to humans, provided that segmentation of the arteries is adapted. This may include fine tuning the vessel-enhancement filter (maximum sphere radius) and extending the number of arterial clusters to four (internal carotid arteries). Also, it should be possible to apply the method to studies with other radiotracers. Depending on the tracer, additional blood samples might be neccessary to determine the plasma input function. If long dynamic PET scans are used (e.g., 18F-fludeoxyglucose (FDG) uptake during approximately 1 h), it would be useful to repeat the TOF-MRA measurement during this time in order to track potential movement of the arteries.
The small deviation between arterial blood sampling and IDIF during steady state indicates that the correction of PVE and PSF is possible with the method presented. The differences in bolus dynamics and, hence, CBF values can be explained by the different sampling locations (hind leg vs. major neck arteries) with differences in delay/dispersion. It will be the topic of further work to test the method on humans with the perspective of replacing invasive blood sampling by an IDIF using simultaneous PET/MRI.
Funding by the Deutsche Forschungsgemeinschaft (German Research Foundation), project SA 669/9-1, is highly appreciated.
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Chen K, Bandy D, Reiman E, Huang SC, Lawson M, Feng D, et al.Noninvasive quantification of the cerebral metabolic rate for glucose using positron emission tomography 18F-fluoro-2-deoxyglucose, the patlak method, and an image-derived input function. J Cereb Blood Flow Metab. 1998; 18:716–23.View ArticlePubMedGoogle Scholar
- Wahl LM, Asselin MC, Nahmias C. Regions of interest in the venous sinuses as input functions for quantitative PET. J Nucl Med. 1999; 40:1666–75.PubMedGoogle Scholar
- Guo H, Renauta R, Chenb K, Reiman E. Clustering huge data sets for parametric PET imaging. BioSystems. 2003; 71:81–92.View ArticlePubMedGoogle Scholar
- Fang YH, Kao T, Liu RS, Wu LC. Estimating the input function non-invasively for FDG-PET quantification with multiple linear regression analysis: simulation and verification with in vivo data. Eur J Nucl Med Mol Imag. 2004; 31:692–702.View ArticleGoogle Scholar
- Liptrot M, Adams KH, Martiny L, Pinborg LH, Lonsdale MN, Olsen NV, et al.Cluster analysis in kinetic modelling of the brain: a noninvasive alternative to arterial sampling. NeuroImage. 2004; 21:483–93.View ArticlePubMedGoogle Scholar
- Naganawa M, Kimura Y, Ishii K, Oda K, Ishiwata K, Matani A. Extraction of a plasma time-activity curve from dynamic brain PET images based on independent component analysis. IEEE Trans Biomed Engin. 2005; 52:201–10.View ArticleGoogle Scholar
- Fang Y-HD, Muzic Jr. RF. Spillover and partial-volume correction for image-derived input functions for small-animal 18F-FDG PET studies. J Nucl Med. 2008; 49:606–14.View ArticlePubMedGoogle Scholar
- Mourik JEM, Lubberink M, Klumpers UM, Comans EF, Lammertsma AA, Boellaard R. Partial volume corrected image derived input functions for dynamic PET brain studies: methodology and validation for 11C-flumazenil. NeuroImage. 2008; 39:1041–50.View ArticlePubMedGoogle Scholar
- Su Y, Shoghi KI. Single-input-dual-output modeling of image-based input function estimation. Mol Imag Biol. 2010; 12:286–94.View ArticleGoogle Scholar
- Lyoo CH, Zanotti-Fregonara P, Zoghbi SS, Liow JS, Xu R, Pike VW, et al.Image-derived input function derived from a supervised clustering algorithm: methodology and validation in a clinical protocol using [11C](R)-rolipram. PLoS ONE. 2014; 9:89101.View ArticleGoogle Scholar
- Simončič U, Zanotti-Fregonara P. Image-derived input function with factor analysis and a-priori information. Nucl Med Comm. 2015; 36:187–93.View ArticleGoogle Scholar
- Litton JE. Input function in PET brain studies using MR-defined arteries. J Comp Asst Tomogr. 1997; 21:907–9.View ArticleGoogle Scholar
- Fung EK, Carson RE. Cerebral blood flow with [15O]water PET studies using an image-derived input function and MR-defined carotid centerlines. Phys Med Biol. 2013; 58:1903–23.PubMed CentralView ArticlePubMedGoogle Scholar
- da Silva N, Herzog H, Weirich C, Tellmann L, Kops ER, Hautzel H, et al.Image-derived input function obtained in a 3TMR-brainPET. Nucl Instr Meth Phys Res A. 2013; 702:22–5.View ArticleGoogle Scholar
- Su Y, Arbelaez AM, Benzinger TLS, Snyder AZ, Vlassenko AG, Mintunand MA, et al.Noninvasive estimation of the arterial input function in positron emission tomography imaging of cerebral blood flow. J Cereb Blood Flow Metab. 2013; 33:115–21.PubMed CentralView ArticlePubMedGoogle Scholar
- Croteau E, Lavallée E, Labbe SM, Hubert L, Pifferi F, Rousseau JA, et al.Image-derived input function in dynamic human PET/CT: methodology and validation with 11C-acetate and 18F-fluorothioheptadecanoic acid in muscle and 18F-fluorodeoxyglucose in brain. Eur J Nucl Med Mol Imag. 2010; 37:1539–50.View ArticleGoogle Scholar
- Jochimsen TH, von Mengershausen M. ODIN—object-oriented development interface for NMR. J Magn Reson. 2004; 170:67–78.View ArticlePubMedGoogle Scholar
- Boltze J, Förschler A, Nitzsche B, Waldmin D, Hoffmann A, Boltze CM, et al.Permanent middle cerebral artery occlusion in sheep: a novel large animal model of focal cerebral ischemia. J Cereb Blood Flow Metab. 2008; 28:1951–64.View ArticlePubMedGoogle Scholar
- Werner P, Saur D, Zeisig V, Ettrich B, Patt M, Sattler B, et al.Simultaneous PET/MRI in stroke: a case series. J Cereb Blood Flow Metab. 2015; 35:1421–5.View ArticlePubMedGoogle Scholar
- Toumoulin C, Boldak C, Dillenseger JL, Coatrieux JL, Rolland Y. Fast detection and characterization of vessels in very large data sets using geometrical moments. IEEE Trans Biomed Engin. 2001; 48:604–6.View ArticleGoogle Scholar
- Hernández Hoyos M, Orłowski P, Piatkowska-Janko E, Bogorodzki P, Orkisz M. Vascular centerline extraction in 3D MR angiograms for phase contrast MRI blood flow measurement. Int J Comp Asst Radiol Surg. 2006; 1:51–61.View ArticleGoogle Scholar
- Sattler B, Jochimsen T, Barthel H, Sommerfeld K, Stumpp P, Hoffmann KT, et al.Physical and organizational provision for installation, regulatory requirements and implementation of a simultaneous hybrid PET/MR-imaging system in an integrated research and clinical setting. Magn Reson Mat Phys Biol Med. 2013; 26:159–71.View ArticleGoogle Scholar
- van Velden FHP, Kloet RW, van Berckel BNM, Wolfensberger SPA, Lammertsma AA, Boellaard R. Comparison of 3D-OP-OSEM and 3D-FBP reconstruction algorithms for high-resolution research tomograph studies: effects of randoms estimation methods. Phys Med Biol. 2008; 53:3217–30.View ArticlePubMedGoogle Scholar
- Rousset OG, Ma Y, Evans AC. Correction for partial volume effects in PET: principle and validation. J Nucl Med. 1998; 39:904–11.PubMedGoogle Scholar
- Alpert NM, Eriksson L, Chang JY, Bergstrom M, Litton JE, Correia JA, et al.Strategy for the measurement of regional cerebral blood flow using short-lived tracers and emission tomography. J Cereb Blood Flow Metab. 1984; 4:28–34.View ArticlePubMedGoogle Scholar
- Meyer E. Simultaneous correction for tracer arrival delay and dispersion in CBF measurements by the H2-15O autoradiographic method and dynamic PET. J Nucl Med. 1989; 30:1069–78.PubMedGoogle Scholar
- Frangi AF, Niessen WJ, Vincken KL, Viergever MA. Multiscale vessel enhancement filtering. Proc Med Imag Comp Comp Asst Interv (MICCAI). 1998; 1496:130–7.Google Scholar