- Meeting abstract
- Open Access
Direct evaluation of MR-derived attenuation correction maps for PET/MR of the mouse myocardium
EJNMMI Physics volume 1, Article number: A85 (2014)
Attenuation correction (AC) must be applied to provide accurate measurements of PET tracer activity concentrations. Due to the limited space available in PET/MR scanners, MR-derived AC (MRAC) is used as a substitute for gold standard transmission source scans . We compared MRAC to transmission scans to evaluate its performance in mouse myocardium studies.
PET SUV values derived for 10 mice  using whole body MRAC maps were compared to those attained using AC maps from a transmission source. 3D FISP was acquired using a 4.7T Bruker BioSpec before the mouse was transferred on a standard Bruker animal bed (with single loop surface coil) to the Cambridge split magnet PET/MR . A 10 minute transmission scan (68Ge) was performed. Emission data was acquired for 45 minutes following ~25MBq 18F-FDG administration.
MRAC comparison Following co-registration using SPMMouse , MR data were forward projected into 3D PET sinograms and thresholded to create an AC map, defined as a single region of tissue with uniform attenuation co-efficient of 0.095cm–1. SUV values were calculated from summed PET images (last 20 minutes) and compared on a voxel by voxel basis between images without AC, with transmission source AC, and with MRAC.
A 22.6 ± 0.9% (mean ± SD) improvement in mouse myocardium SUV values (shown in Figures 1 and 2) was seen by applying transmission AC and a 18.5 ± 0.9% improvement using MRAC, compared to not applying AC. The global attenuation correction over the whole mouse body was 20.7 ± 0.7% using transmission AC and 16.5 ± 1.3% using MRAC. Differences of up to 40% (mean: 30.1 ± 4.4%, range: 27-40%) were seen adjacent to the RF coil (see Figure 3).
A simple, one region MRAC approach provided acceptable AC compared to transmission scanning for myocardial imaging in mice.
Wagenknecht G, et al.: MRI for attenuation correction in PET: methods and challenges. Magn Reson Mater Phy 2013, 26: 99–113. 10.1007/s10334-012-0353-4
Buonincontri G, et al.: PET/MRI assessment of the infarcted mouse heart. NIMA A 2014, 734: 152–5. 0.1016/j.nima.2013/08.066
Lucas AJ, et al.: Development of a combined microPETÂ®-MR system. IEEE Nucl Sci Symp Record 2006, 4: 2345–8. 10.1109/NSSMIC.2006.354384
Sawiak SJ, et al.: MRI reveals brain asymmetry following 6-OHDA lesions in the mouse brain. Proc. ISMRM 2009, 17: 1077. [http://cds.ismrm.org/protected/09MProceedings/files/01077.pdf]
About this article
Cite this article
Evans, E., Buonincontri, G., Hawkes, R.C. et al. Direct evaluation of MR-derived attenuation correction maps for PET/MR of the mouse myocardium. EJNMMI Phys 1, A85 (2014) doi:10.1186/2197-7364-1-S1-A85
- Attenuation Correction
- Transmission Source
- Mouse Myocardium
- Uniform Attenuation
- Voxel Basis