- Meeting abstract
- Open Access
PET motion correction using MR-derived motion parameters
EJNMMI Physics volume 1, Article number: A53 (2014)
With the improving resolution of modern PET scanners, any slight motion during the scan can cause significant blurring and loss of resolution. MRI scanners have the capacity to perform quick successive scans and thus provide a means to track motion during a scan. Hence, with the advent of simultaneous PET-MR scanners, it has become possible to use the MR scanner to track the motion and thereby provide the necessary motion parameters to correct the PET data. Using a suitable segmentation approach a separate MR scan can provide the attenuation map to produce quantitative PET images.
An FDG brain scan was acquired on a Siemens Biograph mMR PET-MR scanner. The MR scan was acquired using the Golden Angle Radial Sparse Parallel (GRASP) sequence , simultaneously with a 5 minute PET scan, while the patient was asked to move his head repetitively from side to side for proof-of-principle purposes. A separate static scan was also acquired prior to the motion scan, to be used as a control. The MR data were divided into a series of 268 images with a frequency of approximately 1 Hz. The motion parameters were derived by performing a rigid (6 degrees-of-freedom) registration of the masked MR images to a chosen reference image. The PET list-mode data were corrected on an event-by-event basis [2, 3]. List-mode maximum likelihood expectation-maximisation (accelerated with ordered subsets ) was used for the reconstruction, incorporating the attenuation correction (as a pre-correction to the data) as well as weighted-average sensitivity  to achieve a quantitative reconstruction.
Motion correction successfully removed almost all motion artefacts, recovered the resolution and allowed for quantitative images to be produced. Techniques to improve upon the coarse sampling of the MR images, such as interpolating between motion data points, will be investigated.
Feng L, Grimm R, Tobias Block K, Chandarana H, Kim S, Xu J, Axel L, Sodickson DK, Otazo R: Golden-angle radial sparse parallel MRI: Combination of compressed sensing, parallel imaging, and golden-angle radial sampling for fast and flexible dynamic volumetric MRI. Magn. Reson. Med. 2013., 00: 10.1002/mrm.24980
Rahmim A, Bloomfield P, Houle S, Lenox M, Michel C, Buckley KR, Ruth TJ, Sossi V: Motion Compensation in Histogram-Mode and List-Mode EM Reconstructions: Beyond the Event-Driven Approach. IEEE Trans. Nucl. Sci. 2004,51(5):2588–2596. 10.1109/TNS.2004.835763
Kyme AZ, Zhou VW, Meikle SR, Baldock C, Fulton RR: Optimised motion tracking for positron emission tomography studies of brain function in awake rats. PLoS One 2011,6(7):e21727. 10.1371/journal.pone.0021727
Hudson HM, Larkin RS: Accelerated image reconstruction using ordered subsets of projection data. IEEE Trans. Med. Imaging 1994, 13: 601–609. 10.1109/42.363108
About this article
Cite this article
Bickell, M., Koesters, T., Boada, F. et al. PET motion correction using MR-derived motion parameters. EJNMMI Phys 1, A53 (2014). https://doi.org/10.1186/2197-7364-1-S1-A53
- Motion Parameter
- Motion Correction
- Siemens Biograph
- Angle Radial
- Coarse Sampling