- Meeting abstract
- Open Access
Highly accelerated Point-Spread Function mapping based on Finite Rate of Innovation for EPI distortion correction
© Nunes and Hajnal; licensee Springer 2014
Published: 29 July 2014
Images were acquired on a 3.0T Philips Achieva: 2.5×2.5×4.0 mm3, matrix 96×95, 95 ks steps with under-sampling retrospectively performed. PSF peak position was estimated using both the fully-sampled (zero-filled by a factor of 1000) and highly-undersampled data.
To determine the PSF peak location, pattern matching was performed. The signal measured at each spatial location was compared to a predicted signal pattern accounting for the ks sampling scheme and the search progressively refined up to the intended precision. EPI images were undistorted as in .
Using the proposed approach the position of the PSF peak can be estimated from a very small number of samples. In the future distortion map estimation could easily be incorporated into standard preparation phases. Making distortion correction of EPI images more practical would facilitate combining functional PET and MR information as well as structural connectivity information from diffusion-weighted MR images.
- Jezzard P, et al.: Correction for geometric distortion in echo planar images from B0 field variations. MRM 1995, 34: 65–73. 10.1002/mrm.1910340111PubMedView ArticleGoogle Scholar
- Robson MD, et al.: Measurement of the point spread function in MRI using constant time imaging. MRM 1997, 38: 733. 10.1002/mrm.1910380509PubMedView ArticleGoogle Scholar
- Zeng H, et al.: Image distortion correction in EPI: Comparison of field mapping with point spread function mapping. MRM 2002, 48: 137–146. 10.1002/mrm.10200PubMedView ArticleGoogle Scholar
- Zaitsev M, et al.: Point spread function mapping with parallel imaging techniques and high acceleration factors: Fast, robust, and flexible method for echo-planar imaging distortion correction. MRM 2004, 52: 1156–66. 10.1002/mrm.20261PubMedView ArticleGoogle Scholar
- In M, et al.: Highly accelerated PSF-mapping for EPI distortion correction with improved fidelity. MagnResonMaterPhy 2012, 25: 183–192.Google Scholar
- Dragonu I, et al.: Accelerated point spread function mapping using signal modeling for accurate echo-planar imaging geometric distortion correction. MRM 2013, 69: 1650–56. 10.1002/mrm.24396PubMedView ArticleGoogle Scholar
- Nunes RG, et al.: ESMRMB. 2013, 30: 145.Google Scholar
- Vetterli M, et al.: Sampling signals with finite rate of innovation. IEEE Trans Sig Process 2002, 50: 1417–28. 10.1109/TSP.2002.1003065View ArticleGoogle Scholar
- Maravic I, et al.: IEEE Trans Sig Process. 2005, 53: 2788–2805.View ArticleGoogle Scholar
- Oh SH, et al.: Distortion correction in EPI at ultra-high-field MRI using PSF mapping with optimal combination of shift detection dimension. MRM 2012, 68: 1239–46. 10.1002/mrm.23317PubMedView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.