- Opinion article
- Open Access
The contribution of Medical Physics to Nuclear Medicine: looking back - a physicist’s perspective
EJNMMI Physics volume 1, Article number: 2 (2014)
This paper is the first in a series of invited perspectives by four pioneers of Nuclear Medicine imaging and physics. A medical physicist and a Nuclear Medicine clinical specialist each take a backward look and a forward look at the contributions of Medical Physics to Nuclear Medicine.
Contributions of Medical Physics are presented from the early discovery of radioactivity, development of first imaging devices, computers and emission tomography to recent development of hybrid imaging.
There is evidence of significant contribution of Medical Physics throughout the development of Nuclear Medicine.
Nuclear Medicine has its basis in ‘the use of radioactive tracers for medical diagnosis, therapy’. Since the inception of the discipline, the adoption of scientific methods and instruments has been central to its development and practice. Even today, the Nuclear Medicine service, at least in the larger tertiary medical centres and teaching hospitals, involves a team of individuals that includes scientists, clinicians, technologists, nurses and ancillary staff. The strength of Nuclear Medicine continues to be its unique sensitivity to picomolar concentrations of radiotracer that can be used to probe underlying physiological, biochemical and molecular processes. The exploitation of this capability continues to rely on joint ingenuity of scientific and clinical specialists.
The origins of Nuclear Medicine arose from the disciplines of both chemistry and physics (notably Marie Sklodowska Curie was awarded Nobel prizes in both fields). Chemistry was central to the discovery and production of radionuclides that are suitable for use in humans, and the development of specific radiolabelled compounds continues to challenge radiochemists and is central to evolving clinical practice. Equally important has been the development of instrumentation and image analysis tools for measurement and depiction of the in vivo radiotracer distribution that has involved a range of scientific disciplines. The focus in this article is on this latter group of scientists, collectively referred to as ‘medical physicists’, but including physicists, engineers, computer scientists, mathematicians and statisticians (including technologically oriented clinicians) who all contribute to the technological development that underpins Nuclear Medicine. A full review of the contributions of Medical Physics to Nuclear Medicine is well beyond the scope of this short article; instead, selected historical examples will be used to highlight the continuing significant clinical impact that medical physicists contribute to the field. Readers are referred to some general texts [1–4] and historical reviews [5–10] that provide more detailed description of technological developments in addition to selected key articles.
Pre 1940: early pioneers
Clearly, the discovery of radioactivity by Henri Becquerel (1896)  and radium by Pierre and Marie Curie (1898) [12, 13] sets the scene for the whole discipline, and as early as 1913, the tracer principle was enunciated by Georg de Hevesy (a chemist)  leading to early use of radionuclides in humans by Blumgart and Weiss  and the first external radiation detectors. The invention of the cyclotron in 1932 by Ernest Lawrence (an engineer) , and subsequently the nuclear reactor, led very quickly to the production of useful radionuclides (e.g. iodine-131)  which were immediately used in early clinical studies . Effectively, Nuclear Medicine was born.
1940 to 1960: from measurement to images
Early inventions of scintillation probes and counters as well as the Geiger counter continue to be used today in a range of applications, although limited in providing any spatial information (see ). They did however facilitate the development of dynamic studies to evaluate function and an early interest in understanding tracer dilution and uptake patterns, techniques that still permeate Nuclear Medicine practice, especially in the research setting. These formed the basis for clinical studies of renal function and cerebral blood flow. The invention of the rectilinear scanner by Benedict Cassen in 1951 , however, provided for the first time an image of the radiotracer distribution. This scanner in combination with in vitro studies and probe-based dynamic measurements provided the tools for early establishment of many Nuclear Medicine centres worldwide (in combination with use of radionuclides for therapy). But probably the invention that has had the highest impact on Nuclear Medicine practice is that of the gamma camera by Hal Anger in 1958 . Detection of positron emission by coincidence detection using probes predated this [22, 23], but early imaging was mainly limited to single photon detection using either a rectilinear scanner or gamma camera.
1960 to 1980: computing and the birth of emission tomography
It took several years for the gamma camera to appear on the commercial market, and in the ensuing years, there was significant development of tomographic devices, initially based on probe systems but subsequently based on use of the basic gamma camera. David Kuhl is credited with development of the first tomographic systems (both emission and transmission) with reconstruction involving optical back projection without use of a computer [24, 25]. It should be remembered that computers were in their infancy at that time and researchers had to resort to innovative alternatives in order to reconstruct even crude images. It was not until the early 1970s  that computer-based reconstruction was introduced, very much influenced by the introduction of X-ray CT and availability of filtered back projection. In the meantime, a number of novel approaches were developed including focal plane (limited angle) tomography and early studies where the patient was rotated in front of a stationary camera . The rotating gamma camera, the basis for most current single photon emission computed tomography (SPECT) systems, was first introduced by John Keyes in 1977 , with commercial demonstration by Ron Jaszczak around the same time .
The development of positron emission tomography (PET) followed a similar course with the earliest dual planar systems developed by Gordon Brownell at Mass General , and the first ring systems built in the early 1970s by Robertson, Yamamoto and colleagues at Brookhaven  and Michael Ter-Pogossian in conjunction with Mike Phelps and Ed Hoffman , who went on to install the first commercial system in 1976 and pioneer further early PET systems at UCLA. As with other sectors of Nuclear Medicine, it was the combination of chemistry (synthesis of FDG) and engineering/physics (development of medical cyclotrons and robust PET cameras) that triggered the widespread use of PET, initially as a research system but ultimately as an important clinical tool.
1980 to 2000: emission tomography comes of age
Both SPECT and PET became more widely available in this period of consolidation with both modalities being demonstrated to have health impact that justified their clinical use. During this period, early deficiencies were recognised and rectified and robust commercial systems developed. In the case of SPECT, this involved adoption of dual detectors and robust rotation gantries rather than the initial counter-balanced single head systems. PET systems benefitted from new technology (block detectors) initially using BGO and later LSO and LYSO. An important contributor to the efficacy of tomographic imaging was the development of iterative reconstruction based on maximum likelihood (MLEM) [33, 34], especially the accelerated form based on ordered subsets (OSEM) , which permitted computationally demanding algorithms to perform in a clinically acceptable time. The recent resurgence of time-of-flight measurement on systems using faster scintillators takes advantage of iterative reconstruction to provide signal-to-noise benefit.
The greater clinical demand for SPECT has also stimulated further novel systems specifically designed for a specific purpose (e.g. cardiac [36–38]). Previous focus on neurological systems proved less popular outside the research setting. These systems have taken advantage of detector technology (e.g. CZT) that has more recently been translated from high-energy physics laboratories, resulting in compact systems that have improved performance compared to the traditional gamma camera. These systems also offer potential for novel acquisition, adapting to the specific patient or study. As with most of the tomographic systems, developments in technology have driven the application, providing increasingly superior image quality and diagnostic capability.
2000 to present: hybrid imaging takes over
Both PET and SPECT benefit from attenuation correction based on direct estimation of the in vivo attenuation by means of transmission measurement [39, 40]. Developments of transmission systems for PET and SPECT were both based on external sources that could provide attenuation maps for correction. However, these were typically of poor quality, with noise in the transmission study propagating so as to deteriorate the emission study. Bruce Hasegawa initially suggested use of CT , leading to the introduction of a ‘low-cost, low-dose’ SPECT/CT in 1999 by GE Healthcare . Subsequent developments have led to a range of fully diagnostic SPECT/CT systems. In the case of PET/CT, this was conceived initially in Geneva by David Townsend and colleagues, then developed at Pittsburgh (in conjunction with Ron Nutt at CTI) [43–45] and ultimately introduced commercially in 2000. The impact of PET/CT has been particularly impressive as the added diagnostic value of the combined modalities has by far exceeded expectations rather than simply providing efficient high-quality transmission measurement for the purpose of attenuation correction [46, 47].
The most recent hybrid imaging combination to be commercially released is PET/MRI, opening further potential for combining information and exploring joint synergies. This is another fine example of how innovative technology deriving outside the medical field (in this case, solid state readout systems that replace the photomultiplier of the conventional detector) can be adapted for benefit (magnetic compatibility, compact design, improved performance). The early developments in preclinical systems, e.g. , rapidly led to the introduction of paired sequential PET/MRI systems before the first demonstration of simultaneous PET/MRI in human neurological studies, quickly followed by whole-body applications [49, 50]. SPECT/MRI is not commercially available at this time, but development is in progress ; whether this proves clinically useful remains to be seen.
The brief history above demonstrates very clearly how the growth of Nuclear Medicine can be attributed, at least in part, to the technological development. This history overlooks the associated research across the breadth of Nuclear Medicine that has led to the current practice. This includes international guidelines on radiation safety and dosimetry, accepted procedures for instrument specification, acceptance testing and quality control, and standards for digital image storage and transfer. The techniques used to guarantee image fidelity that are incorporated in commercial systems have been developed and validated (e.g. uniformity and linearity correction), the software for various image corrections necessary for quantification has evolved and continues to be developed (corrections for attenuation, scatter, resolution and motion), and image analysis software that facilitates quantification, image fusion, kinetic analysis and disease classification represents enormous international effort with significant impact on current clinical capabilities. The ‘medical physicist’ plays an important role as a member of the Nuclear Medicine team, providing scientific support in the clinical setting; equally, he/she continues to innovate and validate techniques that aid in improving the quality of information that can be extracted from medical images as well as the effectiveness of treatment. There are many active university-based research laboratories that continue to focus on the development of medical imaging instrumentation, image reconstruction and image analysis with many internationally recognised researchers contributing to the continuing development. It is impossible in this brief article to give adequate credit to the many individuals who have made major contributions to the field and have greatly influenced the tools that are available to current Nuclear Medicine practice (Figure 1). The many Medical Physics mentors and teachers should also not be overlooked as they have enabled the whole Nuclear Medicine community. The Medical Physics community also provides the manpower that populates the laboratories and support staff of the continually developing vendors.
It is hard to overstate the value of the contribution of medical physicists to Nuclear Medicine. The development of the discipline has been driven by innovation from both scientists and clinicians. This partnership has been important in defining clinically relevant questions and seeking novel but practical solutions. Technology continues to develop (e.g. new detector materials and faster computers), and Medical Physics innovation continues to push the frontiers of what is possible. Effective translation from research laboratories to clinics requires specialised expertise and often lacks specific funding. Likewise, demonstration of clinical utility and impact is an important endeavour that requires dedicated staff or allocation of time, protected from normal clinical support demands. Medical physicists have been central to the development of Nuclear Medicine and must continue this essential work.
Cherry SR, Sorenson JA, Phelps ME: Physics in Nuclear Medicine. USA: Elsevier Health Sciences; 2003.
Lawson RS: The Gamma Camera: A Comprehensive Guide. York, UK: IPEM; 2013.
Khalil MM (Ed): Basic Science of Nuclear Medicine. Heidelberg: Springer; 2011.
Wernick MN, Aarsvold JN (Eds): Emission Tomography: The Fundamentals of SPECT and PET. San Diego, CA: Elsevier; 2004.
Morris RR: Society of Nuclear Medicine, 1954–2004: 50 Years of Excellence. Reston, VA: SNM; 2004.
Webb S: From the Watching of Shadows: The Origins of Radiological Tomography. Bristol, UK: Adam Hilger; 1990.
Hutton BF: The origins of SPECT and SPECT/CT. Eur J Nucl Med Mol Imaging 2013. doi:10.1007/s00259–013–2606–5 doi:10.1007/s00259-013-2606-5
Nutt R: The history of positron emission tomography. Mol Imaging Biol 2002, 4: 11–26. 10.1016/S1095-0397(00)00051-0
Jones T: Historical development of functional in vivo studies using positron-emitting tracers. In Positron Emission Tomography: Basic Sciences. Edited by: Bailey DL, Townsend DW, Valk PE, Maisey MN. London, UK: Springer; 2005:3–38.
Muehllehner G, Karp JS: Positron emission tomography. Phys Med Biol 2006, 51: R117-R137. 10.1088/0031-9155/51/13/R08
Becquerel H: Sur les radiations invisibles emises par les corps phosphorescents. Comptes rendus de l'Academie des Sciences, Paris 1896, 122: 501–503.
Curie P, Curie M, Bemont G: Sur une nouvelle substance fortement radio-active contenue dans la pechblende. Comptes Rendus des l’Academie des Sciences, Paris 1898, 127: 1215–1217.
“Marie Curie - Nobel Lecture: Radium and the New Concepts in Chemistry” Nobelprize.org Nobel Media AB 2013. [http://www.nobelprize.org/nobel_prizes/chemistry/laureates/1911/marie-curie-lecture.html] Nobel Media AB 2013.
de Hevesy G: Radioelements as tracers in chemistry and physics. Chem News 1913, 108: 166.
Blumgart HL, Weiss S: Studies on the velocity of blood flow. J Clin Invest 1927, 4: 15–31. 10.1172/JCI100110
Lawrence EO, Livingstone MS: The production of high-speed light ions without the use of high voltages. Phys Rev 1932, 40: 19–30. 10.1103/PhysRev.40.19
Livingwood JJ, Seaborg GT: Radioactive isotopes of iodine. Phys Rev 1938, 54: 775–782. 10.1103/PhysRev.54.775
Hamilton JG, Soley MH: Studies in iodine metabolism by the use of a new radioactive isotope of Iodine. Amer Physiol Soc 1939, 127: 557072.
Knoll GF: Radiation Detection and Measurement (4th ed). US: John Wiley; 2010.
Cassen B, Curtis L, Reed C, Libby R: Instrumentation for 131I use in medical studies. Nucleonics 1951, 9: 46–50.
Anger HO: Scintillation camera. Rev Sci Instr 1958, 29: 27–33. 10.1063/1.1715998
Wrenn FR, Good ML, Handler P: The use of positron-emitting radioisotopes for the localization of brain tumours. Science 1951, 113: 525–527. 10.1126/science.113.2940.525
Sweet WH: The use of nuclear disintegrations in diagnosis and treatment of brain tumours. New Eng J Med 1951, 245: 875–878. 10.1056/NEJM195112062452301
Kuhl DE, Edwards RQ: Image separation radioisotope scanning. Radiology 1963, 80: 653–662.
Kuhl DE, Hale J, Eaton WL: Transmission scanning: a useful adjunct to conventional emission scanning for accurately keying isotope deposition to radiographic anatomy. Radiology 1966, 87: 278–284.
Muellehner G, Wetzel RA: Section imaging by computer calculation. J Nucl Med 1971, 12: 76–85.
Anger HO, Price DC, Yost PE: Transverse section tomography with the scintillation camera. J Nucl Med 1967, 8: 314.
Keyes JW, Orlandea N, Heetderks WJ, Leonard PF, Rogers WL: The Humongotron – a scintillation camera transaxial tomography. J Nucl Med 1977, 18: 381–387.
Jaszczak RJ, Murphy PH, Huard D, Burdine JA: Radionuclide emission computed tomography of the head with 99mTc and a scintillation camera. J Nucl Med 1977, 18: 373–380.
Brownell GL, Burnham CA, Wilensky S, Arnonow S, Kazemi H, Strieder D: New Developments in Positron Scintigraphy and the Application of Cyclotron-Produced Positron Emitters. Medical Radioisotope Scintigraphy. Vienna: IAEA; 1969:163–176.
Robertson JS, Marr RB, Rosenblum M, Radeka V, Yamamoto YL: 32-crystal positron transverse section detector. In Tomographic Imaging in Nuclear Medicine. Edited by: Freedman GS. New York: Society of Nuclear Medicine; 1973:142–153.
Phelps ME, Hoffman EJ, Mullani NA, Ter Pogossian MM: Application of annihilation coincidence detection of transaxial reconstruction tomography. J Nucl Med 1975, 16: 210–215.
Shepp LA, Vardi Y: Maximum likelihood reconstruction in positron emission tomography. IEEE Trans Med Imaging 1982, 1: 113–122.
Lange K, Carson R: EM Reconstruction algorithms for emission and transmission tomography. J Comput Assist Tomogr 1984, 8: 306–316.
Hudson HM, Larkin RS: Accelerated image reconstruction using ordered subsets of projection data. IEEE Trans Med Imaging 1994, 13: 601–609. 10.1109/42.363108
Garcia EV, Faber TL, Esteves FP: Cardiac dedicated ultrafast SPECT cameras: new designs and clinical implications. J Nucl Med 2011, 52: 210–217. 10.2967/jnumed.110.081323
Patton JA, Slomka PJ, Germano G, Berman DS: Recent technological advances in nuclear cardiology. J Nucl Cardiol 2007, 14: 555–565. 10.1016/j.nuclcard.2007.04.022
Hutton BF: Developments in cardiac-specific SPECT imaging. Q J Nucl Med 2012, 56: 221–229.
Bailey DL: Transmission scanning in emission tomography. Eur J Nucl Med 1998, 25: 774–787. 10.1007/s002590050282
Kinahan PE, Hasegawa BH, Beyer T: X-ray-based attenuation correction for positron emission tomography/computed tomography scanners. Seminars Nucl Med 2003, 33: 166–179. 10.1053/snuc.2003.127307
Hasegawa BH, Gingold EL, Reilly SM, Liew SC, Cann CE: Description of a simultaneous emission-transmission CT system. Proc SPIE 1990, 1231: 50–60. 10.1117/12.18783
Patton JA, Delbeke D, Sandler MP: Image fusion using an integrated, dual-head coincidence camera with X-ray tube-based attenuation maps. J Nucl Med 2000, 41: 1364–1368.
Townsend DW, Beyer T, Kinahan PE, Brun T, Roddy R, Nutt R, Byars LG: The SMART scanner: a combined PET/CT tomograph for clinical oncology. Proc IEEE Nucl Sci Symp Med Imaging Conf 1998, 2: 1170–1174.
Beyer T, Townsend DW, Brun T, Kinahan PE, Charron M, Roddy R, Jerin J, Young J, Byars L, Nutt R: A combined PET/CT scanner for clinical oncology. J Nucl Med 2000, 41: 1369–1379.
Townsend DW: Multimodality imaging of structure and function. Phys Med Biol 2008, 53: R1-R39. 10.1088/0031-9155/53/4/R01
Patton JA, Townsend DW, Hutton BF: Hybrid imaging technology: from dreams and vision to clinical devices. Seminars Nucl Med 2009, 39: 247–263. 10.1053/j.semnuclmed.2009.03.005
Cherry SR: Multimodality in vivo imaging systems: twice the power or double the trouble? Annu Rev Biomed Eng 2006, 8: 35–62. 10.1146/annurev.bioeng.8.061505.095728
Shao Y, Cherry SR, Farahani K, Meadors K, Siegel S, Silverman RW, Marsden PK: Simultaneous PET and MR imaging. Phys Med Biol 1997, 42: 1965–1970. 10.1088/0031-9155/42/10/010
Judenhofer MS, Wehrl HF, Newport DF, Catana C, Siegel SB, Becker M, Thielscher A, Kneilling M, Lichy MP, Eichner M, Klingel K, Reischl G, Widmaier S, Röcken M, Nutt RE, Machulla H-J, Uludag K, Cherry SR, Claussen CD, Pichler BJ: Simultaneous PET-MRI: a new approach for functional and morphological imaging. Nat Med 2008, 14: 459–465. 10.1038/nm1700
Drzezga A, Souvatzoglou M, Eiber M, Beer AJ, Fürst S, Martinez-Möller A, Nekolla SG, Ziegler S, Ganter C, Rummenev EJ, Schwaiger M: First clinical experience with integrated whole-body PET/MR: comparison to PET/CT in patients with oncologic diagnoses. J Nucl Med 2012, 53: 845–855. 10.2967/jnumed.111.098608
Busca P, Fiorini C, Butt AD, Occhipinti M, Peloso R, Quaglia R, Schembari F, Trigilio Pb, Nemeth G, Major P, Erlandsson K, Hutton BF: INSERT: development of a new multi-modality SPECT/MRI system for preclinical and clinical use. Nucl Instr Methods Phys Res A 2013. in press in press
The Institute of Nuclear Medicine benefits from support from the National Institute for Health Research University College London Hospitals Biomedical Research Centre.
The author declares that he has no competing interests.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
About this article
Cite this article
Hutton, B.F. The contribution of Medical Physics to Nuclear Medicine: looking back - a physicist’s perspective. EJNMMI Phys 1, 2 (2014) doi:10.1186/2197-7364-1-2
- Nuclear Medicine