Initial evaluation of a practical PET respiratory motion correction method in clinical simultaneous PET/MRI
© Manber et al; licensee Springer 2014
- Published: 29 July 2014
- Respiratory Signal
- Uncorrected Image
- Respiratory Motion Correction
- Multiple Liver Lesion
- Motion Correction Method
Respiratory motion during PET acquisitions can cause image artefacts, with sharpness and tracer quantification adversely affected due to count ‘smearing’. Motion correction by registration of PET gates becomes increasingly difficult with shorter scan times and less counts. The advent of simultaneous PET/MRI scanners allows the use of high spatial resolution MRI to capture motion states during respiration [1, 2]. In this work, we use a respiratory signal derived from the PET list-mode data [3, 4], with no requirement for an external device or MR sequence modifications.
In the liver case we see a decrease in tumor ‘smearing’ after MRI model-based correction (Figure 3). Other areas of high activity in the liver, only marginally visible in the uncorrected image, become apparent in the motion-corrected image. Average intensity increase over the 3 lesions is 11%, while increase in intensity in the cardiac wall in the cardiac patient is 10%.
Siemens/UCL IMPACT studentship, UK EPSRC (EP/K005278/1) and the NIHR-funded UCH Biomedical Research Centre.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (https://creativecommons.org/licenses/by/4.0), which permits use, duplication, adaptation, distribution, and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Chun SY, et al.: MRI-based nonrigid motion correction in simultaneous PET/MRI. J Nucl Med 2012,53(1):1284–91.PubMed CentralPubMedView ArticleGoogle Scholar
- King AP, et al.: Thoracic respiratory motion estimation from MRI using a statistical model and a 2-D image navigator. Medical Image Analysis 2012, 16: 252–264. 10.1016/j.media.2011.08.003PubMedView ArticleGoogle Scholar
- Thielemans K, et al.: Device-less gating for PET/CT using PCA. Nuclear Science Symposium and Medical Imaging Conference (NSS/MIC) 2011, 3904–3910.Google Scholar
- Manber , et al.: ISMRM Conf. Proc. 2014.Google Scholar
- Polycarpou I, et al.: Analysis and comparison of two methods for motion correction in PET imaging. Medical Physics 2012, 39: 6374–6483.View ArticleGoogle Scholar
- Thielemans K, et al.: STIR: software for tomographic image reconstruction release 2. Phys Med Biol 2012,57(4):867–883. 10.1088/0031-9155/57/4/867PubMedView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.