A CZT-based blood counter for quantitative molecular imaging
© The Author(s) 2017
Received: 23 December 2016
Accepted: 2 May 2017
Published: 2 June 2017
Robust quantitative analysis in positron emission tomography (PET) and in single-photon emission computed tomography (SPECT) typically requires the time-activity curve as an input function for the pharmacokinetic modeling of tracer uptake. For this purpose, a new automated tool for the determination of blood activity as a function of time is presented.
The device, compact enough to be used on the patient bed, relies on a peristaltic pump for continuous blood withdrawal at user-defined rates. Gamma detection is based on a 20 × 20 × 15 mm3 cadmium zinc telluride (CZT) detector, read by custom-made electronics and a field-programmable gate array-based signal processing unit. A graphical user interface (GUI) allows users to select parameters and easily perform acquisitions.
This paper presents the overall design of the device as well as the results related to the detector performance in terms of stability, sensitivity and energy resolution. Results from a patient study are also reported. The device achieved a sensitivity of 7.1 cps/(kBq/mL) and a minimum detectable activity of 2.5 kBq/ml for 18F. The gamma counter also demonstrated an excellent stability with a deviation in count rates inferior to 0.05% over 6 h. An energy resolution of 8% was achieved at 662 keV.
The patient study was conclusive and demonstrated that the compact gamma blood counter developed has the sensitivity and the stability required to conduct quantitative molecular imaging studies in PET and SPECT.
KeywordsCZT Gamma counter Blood activity Molecular imaging PET scan TAC
Positron emission tomography (PET) and single-photon emission computed tomography (SPECT) are well-established molecular imaging modalities used in many fields of the biomedical sciences. They allow in vivo investigations of biological processes at the molecular level and provide valuable information on the onset and progression of diseases. The images obtained are based on a measurable number of nuclear disintegrations and, as such, are inherently quantitative. However, the quantitative nature of these modalities is usually dismissed, largely because tools and methods dedicated to quantitative imaging are lacking. Accurate quantification in PET and SPECT typically requires frequent assessments of blood activity, for example through manual sampling and subsequent measurements in a well counter. These methods however can be inaccurate and error-prone, and they expose the personnel to radiation and blood pathogen health hazards. The manual sampling methods might also be difficult to apply to tracers with a fast uptake.
Image-derived TACs are also possible and were investigated by several groups (see e.g. [1, 2] and references therein). These methods are less invasive, but they are not suitable for all anatomical sites, acquisition protocols and radiopharmaceuticals. They typically require the presence of a large blood vessel in the field-of-view (FOV), which is not always possible for dynamic acquisitions where the patient bed stays stationary over a particular FOV. Some image-based TAC also require corrections and sometimes suffer from methodological problems that hinder their widespread use . Zanotti-Fregonara et al. for instance found that none out of eight image-based methods for TAC were reliable to estimate cerebral metabolic rate of glucose without resorting to blood sampling [2, 3]. Often, blood sampling is inevitable (to measure plasma vs whole blood TAC, for example) and remains the gold standard in all cases. Consequently, automated blood counters were developed for these reasons.
The majority of blood counters use a photomultiplier tube (PMT) coupled to a scintillation crystal such as BGO [4–8] or GSO [9, 10] to detect gammas. One device was build with an avalanche photodiode coupled to a LSO crystal . This combination has the advantage of being compact compared to PMT-based solutions but suffers from a rather large temperature dependence. Other devices are based on PIN photodiodes and detect charged particles rather than gammas . This design allows for compact devices but might suffer from low sensitivity as the tubing used can stop a significant fraction of electrons and positrons emitted from the blood. A microfluidic approach was recently used to overcome this problem .
In order to build a robust, sensitive and compact device, the cadmium zinc telluride (CZT) detector technology was used in this work [14, 15]. CZT detectors are known for their high energy resolution and stable operation (no temperature dependence). Although CZT will convert less gammas than BGO or GSO for a given volume, it has the potential advantage of requiring less shielding than PMT-based solutions and offers a better energy resolution. These advantages can result in more compact devices that can potentially improve quantification efforts in multi-isotope studies.
The objective of this work is to demonstrate the advantages of a CZT-based blood counter for PET and SPECT quantification. A compact prototype was designed and built to be used in a clinical setting. This paper will describe the main elements of the system and will present results regarding the detection sensitivity, the stability in time, the background noise and the minimum detectable activity (MDA). First results from a patient study are also reported.
General device features
The total amount of blood withdrawn from a patient should be kept as low as possible. McGuill et al. suggested to withdraw no more than 7.5% of the total blood volume . It is therefore important to adjust the withdrawal parameters to the pharmacokinetic behaviour of a given radiopharmaceutical. The device built allows pre-programmed, variable pump rates in order to minimise the total amount of blood withdrawn for a given study while capturing the dynamics of uptake. Everett et al. have reported that in a 924 patients PET study, taking 117 to 137 mL of blood by arterial cannulation led to a single case of adverse effect (thrombotic occlusion) and concluded that the practise was safe, even if the catheter was in place for 5 h on average . Zanotti-Fregonara et al. came to the same conclusions from their experience with more than 3000 patients .
The device must be located as close to the patient as possible to reduce activity dispersion along the catheter. Therefore, a design as compact as possible was sought for the device. The weight of the prototype was approximately 10 kg, including shielding. This allows the use of the device directly on the patient bed, thereby minimising the length of catheters required and reducing the effect of diffusion that affects the time resolution. The activity dispersion along the catheter can be estimated and corrected, typically by monoexponential deconvolution [18, 19] or step function calibration , but this was not implemented yet for the prototype device presented here. An access to radial blood with short catheters is more complicated in the case of brain studies where arms usually rest alongside the patient. In these cases, the system can be positioned at the feet of the patient with a longer catheter, at the expense of larger dispersion. For head studies with one or two bed positions, another possibility is to place the device on a cart beside the patient with a shorter access to the arm.
The detector assembly was shielded with 25 to 35 mm of 97% pure tungsten. A custom-made plastic container was 3D-printed and filled with tungsten cubes, as shown in Fig. 3. The container, which has a slit for the catheter, ensures reproducible positioning. A separate 3D-printed piece (not shown) slides in the slit and maintains the catheter in place. The length of the catheter exposed to the CZT detector is 29 mm. The detector was not shielded where the cables pass; this area was pointing towards the ceiling in the clinical experiments as shown in Fig. 8. In all cases, there was no direct unshielded line of sight between the CZT crystal and the main source of background radiation (the patient).
A field-programmable gate array (FPGA)-based (Cyclone V, Altera, CA, USA) circuit was designed to control the acquisition and the different subsystems of the blood counter. The FPGA chip allows a convenient handling of signals and components through the Nios II processor. The signal workflow is illustrated in Fig. 4. The analog signal from the detector-preamplifier assembly is digitised by a “free-running” quad, 14-bit analog-to-digital converter (ADC, AD9253, Analog Devices, MA, USA) and then routed to the FPGA where programmable filters and thresholds (shaping) are applied. More specifically, the two signals from the virtual coplanar detector are combined (weighted subtraction) on the FPGA to compensate for charge trapping, as described by Luke et al. [23, 26]. A thorough characterisation of the detector, not reported here, allowed to set optimal operation parameters of this anode configuration. A trapezoidal filter was applied on the resulting signal, and the maximum was extracted. Counts exceeding 70 keV were stored in memory while the FPGA is waiting for calls by the host program via a USB connection.
Graphical user interface
The bottom part of the GUI shows either a graph of the activity as a function of time or an energy histogram of detected events, depending on which tab is selected. Four markers can be positioned to define two energy windows, allowing for example dual-isotope studies. The energy windows selected are applied to the activity displayed in the associated tab.
The GUI has a user mode for regular use and a superuser mode for development. The superuser mode allows a manual control of the device and real-time programming of pump rates, motors and data processing parameters. The user mode is meant to be used in a clinical setting and therefore allows the definition and use of pre-programmed acquisition sequences. For safety reasons, it is possible to bypass a sequence with manual control of the peristaltic pump or the detector.
For all acquisitions, pulse height histograms with 256 bins were obtained. The histograms were energy-calibrated with a 137Cs source of 32.7 kBq, and the energy threshold was set to 110 keV for all acquisitions. An estimation of energy resolution was obtained by fitting a Gaussian on the photopeak (662 keV) of the energy spectrum.
Stability over time and catheter positioning reproducibility
The first test over the 6 h acquisition was analysed by fitting the data to a linear function. A χ 2 test was used to verify that the data was Poisson-distributed, as expected.
In the second experiment, an analysis of variance (ANOVA) was performed to determine if the 19 acquisitions over 3 weeks belonged to a distribution with identical parameters (mean and variance).
Another experiment was conducted to verify that the catheter can be positioned in a reproducible manner each time. For a catheter filled with FDG, the catheter was removed and repositioned 20 times and counting was performed for 30 s with a counting time per sample of 1 s. Counting rates were decay-corrected, and an ANOVA was performed to assess reproducibility of catheter positioning. All statistical tests were performed with R (version 3.2.1).
Minimum detectable activity
The minimum detectable activity (MDA) is a crucial characteristic of the device; the counter must detect a small number of counts per second in the relatively high background of a PET scan room. This can be achieved through adequate shielding, coincidence counting or a combination of both. In this work, shielding only was used but the acquisition system of the device was designed with coincidence counting as an optional feature. Typically, the blood activity is lower than 500 kBq/ml at the maximum of the input function for fluorodeoxyglucose (FDG) .
where T is the counting time per sample, N B is the number of background counts recorded during T, f is a factor of radiation yield per disintegration (f = 0.967 here) and s is the sensitivity of the detector as obtained by calibration. The sensitivity—the ratio of recorded count rate and activity concentration—was obtained with a 1.58-mm-inner-diameter catheter filled with 52.5 kBq/ml of FDG. The counting time per sample T used for MDA determination was 3 s, but it can be adjusted between 1 and 30 s to optimise the MDA as a function of background and activity level in the catheter. Figure 7 shows the prototype with FDG circulating in a catheter.
Background counts were measured in a realistic environment, i.e. in a PET scanner room at 1 m from a patient injected with 275 MBq of FDG 40 min prior to the experiment. The background count rate was averaged over a 3 min acquisition. The experiment was repeated with and without the tungsten shielding to estimate its efficacy.
To evaluate the activity dispersion in the tubing used, a step function study was conducted. A three-way valve was added at the end of the tubing (inner diameter of 1.58 mm) in a setup similar to the one used by Munk et al. . Two vials were connected to the valve, one filled with water and the other with a mixture of water and FDG. Measurements were performed at two pump rates (2 and 4.5 ml min −1) and two tubing lengths (80 and 45 cm). The rising part of the step was modelled by an exponential function, f(t) = A(1− exp−t/τ) for each case .
The use of the device in real conditions is essential to verify that it meets clinical requirements and workflows. For this purpose, the device was tested in a clinical setting with prostate cancer patients undergoing dynamic 18F-fluoromethylcholine (FCH) PET studies. As shown in Fig. 8, the patient was in supine position with his arms positioned above his head. Two venous accesses were installed, one in each arm. FCH was injected in the left arm while blood was withdrawn from the right arm (18 gauge needle). Tubing of 76 cm with an inner diameter (ID) of 2.54 mm was connected between the patient and a stopcock. The stopcock interfaced a saline syringe and the tubing (ID of 1.58 mm) going to the P625 pump, the detector module and then the waste container. The blood withdrawal rate was set to 2 ml/min for the acquisition. The patient was injected with a standard activity of 4 MBq/kg for a total of 355 MBq. The injection was performed within 3 s. The pump was started approximately 1 min before the beginning of the acquisition, defined as the time where the FCH is injected while the PET acquisition and the gamma counting are started simultaneously. The dynamic PET scan with a field-of-view centred on the prostate lasted 600 s. To compensate for the transport delay of the blood in the tubing and to extend gamma counting time, 125 s were added to the gamma counting acquisition.
Stability over time and catheter positioning reproducibility
The data is Poisson-distributed as expected, with a reduced χ 2 value of 1.02.
Minimum detectable activity and energy resolution
The sensitivity s obtained through calibration was 7.1 cps/(kBq/mL) for 18F. In the PET scanner room at 1 m from an FDG-injected patient, the measured background counts were 40 cps and 1500 cps with and without tungsten shielding respectively. For a 3 s sampling time, this yields a MDA of 2.5 kBq/ml with the tungsten shielding, which provides approximately a 30-fold reduction in background counts. The measured energy resolution of the detector was 8% at 662 keV.
The results show that the device fulfills compactness, stability and sensitivity requirements for a typical usage in molecular imaging. The measured sensitivity of 7.1 cps/(kBq/mL) translates into a MDA of 2.5 kBq/ml in a PET scanner room, an environment with a high background. This is well below the 500 kBq ml −1 concentration typically encountered in blood for FDG studies . The detector system count rate behaviour was shown to be linear below 500 kBq ml −1, while obeying a non-paralysable model up to 2.5 MBq ml −1 where dead time accounts for a 10% count loss .
This MDA value corresponds to a sampling time of 3 s, and it can be lowered by a factor 3.2 by increasing the sampling time to 30 s as the activity level decreases in the blood. The prototype allows such sequence programming through its GUI.
The device demonstrated high stability with a drift in count rates lower than 0.05% over 6 h. Random checks over 3 weeks also suggested high reliability. This robustness is a typical advantage of solid-state detectors over scintillators/PMT assemblies that might require periodic recalibration.
Comparison of real-time blood counters
Shielding (mm), material
Sensitivity (18F) (cps/(kBq/ml))
MDA (18F) (kBq/ml)
Absolute efficiency (%)
20 × 20 × 15
10.5% at 511 keV
8% at 662 keV
Eriksson et al. 
45 × 45 × 45
Votaw and Shulman 
50 × 25 × 25
6.9% at 511 keV
Boellaard et al. 
D = 60 L = 60
30% of 511 keV
Kudomi et al. 
20 × 20 × 12.9
6.2% at 511 keV
11% at 511 keV
Twilite II (swisstrace)
Laymon et al. 
D = 25 L = 25
19% at 511 keV
Breuer et al. 
50 × 40 × 30
18.5% at 511 keV
22% at 511 keV
Yamamoto et al. 
10 × 20 × 0.5
18.5% at 511 keV
Convert et al. 
30 × 3
Convert et al. 
16 × 2
Roehrbacher et al. 
D = 25 L = 25
Although energy resolution is not a critical feature for blood counting in PET, the CZT detector used here achieved a relatively good value of 8% at 662 keV, better than any crystal-based solution. This energy resolution let efficient counting in dual-isotope studies.
A single detector module was used in the experiments reported here. This makes the device potentially very sensitive to background radiation. In this regard, the acquisition card and associated firmware were designed to accommodate a second detector module and to perform coincidence counting or photopeak summation . This could further reduce the size and weight of the device as less shielding would be required to achieve the same MDA. Further modelling efforts will be conducted to devise the optimal combination of acquisition mode and shielding requirements to make the detector immune to background variations.
It is important to note that positrons have a non-null probability of reaching the CZT crystal, especially for emitters having an E max larger than 1.2 MeV such as 15O (E max = 1.735 MeV) in the geometry proposed here. Positrons with a lower energy will most likely be absorbed by the material located between the emission position and the CZT crystal (1 mm Al, 1.2 mm plastic and 0.8 mm air in our case). Positron events in CZT can lead to different outcomes, one of which is an energy deposition larger than 511 keV (if one or two annihilation photon interact afterwards in CZT). The other possibility, where both photons escape, produce an event which cannot be differentiated from a gamma interaction and contribute to the system deadtime. In all cases, the effect is systematic for a given geometry and isotope and is easily handled with calibration. Therefore, no problems are anticipated for the use of positron emitters other than 18F assuming proper calibration.
Results obtained in this study demonstrate the feasibility of using a CZT-based detector to obtain the input function for PET or SPECT pharmacokinetic modeling. The large CZT detector used here (20 × 20 × 15 mm3) has the advantage of converting a relatively large fraction of incident gamma at 511 keV. Furthermore, the detector module has a compact design requiring less shielding than PMT-based solutions, provides a better energy resolution than crystal-based solutions and can be operated at room temperature. The MDA of the device was 2.5 kBq/ml for 3 s sampling duration, and it can be improved by a factor 3.2 by increasing the sampling time to 30 s for low activity measurements. The compact device was shown to be stable in time and robustness. The measurement of a TAC in a PET study confirms that the device is adequate for use in a clinical setting. Future work will include the development of an automatic blood packaging system so that additional biochemistry and calibration tests can be performed easily. This is especially important for radiopharmaceuticals requiring metabolite corrections.
The authors would like to thank Jean-Michel Guay and Yanik Landry-Ducharme for the insightful comments and suggestion on the design of the device. This work was financed by the Collaborative Health Research Projects program (grant CHRPJ 385866-10), a joint program of the Natural Sciences and Engineering Research Council of Canada (NSERC) and the Canadian Institutes of Health Research (CIHR).
RE, JPM, LAH and PD designed the device. RE, JPM, AF, LAH, LC, JMB and PD participated in the critical revision of the manuscript. RE and LC did the measurements. RE drafted the manuscript and carried out the data analysis and interpretation. JMB supervised the patient study, and RE and AF operated the device. PD approved the final content. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Consent for publication
The informed consent to report the individual patient data was obtained from all participants included in the study.
Ethics approval and consent to participate
All procedures performed in studies involving human participants were in accordance with the ethical standards of the Ethical Review Committee of CHU de Québec–Université Laval. Informed consent was obtained from all individual participants included in the study.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Zanotti-Fregonara P, Chen K, Liow J-S, Fujita M, Innis RB. Image-derived input function for brain PET studies: many challenges and few opportunities. J Cereb Blood Flow Metab. 2011; 31(10):1986–98. http://dx.doi.org/10.1038/jcbfm.2011.107.
- Su Y, Arbelaez AM, Benzinger TL, Snyder AZ, Vlassenko AG, Mintun MA, Raichle ME. Noninvasive estimation of the arterial input function in positron emission tomography imaging of cerebral blood flow. J Cereb Blood Flow Metab. 2013; 33(1):115–21. http://jcb.sagepub.com/content/33/1/115.abstract.
- Zanotti-Fregonara P, Fadaili EM, Maroy R, Comtat C, Souloumiac A, Jan S, Ribeiro M-J, Gaura V, Bar-Hen A, Trebossen R. Comparison of eight methods for the estimation of the image-derived input function in dynamic [18F]-FDG PET human brain studies. J Cereb Blood Flow Metab. 2009; 29(11):1825–35. http://jcb.sagepub.com/content/29/11/1825.abstract.
- Eriksson L, Holte S, Bohm C, Kesselberg M, Hovander B. Automated blood sampling systems for positron emission tomography. IEEE Trans Nucl Sci. 1988; 35(1):703–7. iSSN 0018-9499.View ArticleGoogle Scholar
- Eriksson L, Ingvar M, Rosenqvist G, Stone-Elander S, Ekdahl T, Kappel P. Characteristics of a new automated blood sampling system for positron emission tomography. IEEE Trans Nucl Sci. 1995; 42(4):1007–11. iSSN 0018-9499.View ArticleGoogle Scholar
- Votaw JR, Shulman SD. Performance evaluation of the pico-count flow-through detector for use in cerebral blood flow PET studies. J Nucl Med. 1998; 39(3):509–15. http://jnm.snmjournals.org/cgi/content/abstract/39/3/509.
- Boellaard R, van Lingen A, van Balen SC, Hoving BG, Lammertsma A. Characteristics of a new fully programmable blood sampling device for monitoring blood radioactivity during PET. Eur J Nucl Med Mol Imaging. 2001; 28(1):81–9. http://dx.doi.org/10.1007/s002590000405.
- Laymon CM, Shin D, Carney JP, Ruszkiewicz J, Altenburger D, Becker C, Lopresti BJ, Mason N, Mountz J, Price J, Schavey R, Mathis CA. Evaluation of a commercial radiochromatography module as an arterial blood activity monitor. Phys Med Biol. 2008; 53(2):339–51. http://stacks.iop.org/0031-9155/53/339.
- Kudomi N, Choi E, Yamamoto S, Watabe H, Kim KM, Shidahara M, Ogawa M, Teramoto N, Sakamoto E, Iida H. Development of a GSO detector assembly for a continuous blood sampling system. IEEE Trans Nucl Sci. 2003; 50(1):70–3. iSSN 0018-9499.View ArticleGoogle Scholar
- Yamamoto S, Imaizumi M, Shimosegawa E, Kanai Y, Sakamoto Y, Minato K, Shimizu K, Senda M, Hatazawa J. A compact and high sensitivity positron detector using dual-layer thin GSO scintillators for a small animal PET blood sampling system. Phys Med Biol. 2010; 55(13):3813. http://stacks.iop.org/0031-9155/55/i=13/a=016.
- Breuer J, Grazioso R, Zhang N, Schmand M, Wienhard K. Evaluation of an MR-compatible blood sampler for PET. Phys Med Biol. 2010; 55(19):5883. http://stacks.iop.org/0031-9155/55/i=19/a=017.
- Convert L, Morin-Brassard G, Cadorette J, Rouleau D, Croteau E, Archambault M, Fontaine R, Lecomte R. A microvolumetric β blood counter for pharmacokinetic PET studies in small animals. IEEE Trans Nucl Sci. 2007; 54:173–80.View ArticleGoogle Scholar
- Convert L, Lebel R, Gascon S, Fontaine R, Pratte J-F, Charette P, Aimez V, Lecomte R. Real-time microfluidic blood-counting system for PET and SPECT preclinical pharmacokinetic studies. J Nucl Med. 2016; 57(9):1460–6. http://jnm.snmjournals.org/content/57/9/1460.abstract.
- Espagnet R, Martin J-P, Hamel L-A, Després P. A CdZnTe-based automated blood counter for quantitative molecular imaging In: Jaffray D, editor. World Congress on Medical Physics and Biomedical Engineering, Vol. 51 of IFMBE Proceedings. Toronto: Springer International Publishing: 2015. p. 1338–42.Google Scholar
- Espagnet R, Frezza A, Martin J-P, Hamel L-A, Després P. Conception and characterization of a virtual coplanar grid for a 11x11 pixelated CZT detector. Nucl Instrum Methods Phys Res Sect A Accelerators Spectrometers Detectors Assoc Equip. 2017; 860:62–9. ISSN 0168-9002. http://www.sciencedirect.com/science/article/pii/S0168900217303595.
- McGuill MW, Rowan AN. Biological effects of blood loss: implications for sampling volumes and techniques. ILAR J. 1989; 31(4):5–20. http://ilarjournal.oxfordjournals.org/content/31/4/5.short.
- Everett BA, Oquendo MA, Abi-Dargham A, Nobler MS, Devanand DP, Lisanby SH, Mann JJ, Parsey RV. Safety of radial arterial catheterization in PET research subjects. J Nucl Med. 2009; 50(10):1742. http://jnm.snmjournals.org/content/50/10/1742.short.
- Meyer E. Simultaneous correction for tracer arrival delay and dispersion in CBF measurements by the H215O Autoradiographic Method and Dynamic PET. J Nucl Med. 1989; 30(6):1069–78. http://jnm.snmjournals.org/content/30/6/1069.short.
- Iida H, Kanno I, Miura S, Murakami M, Takahashi K, Uemura K. Error analysis of a quantitative cerebral blood flow measurement using H215O autoradiography and positron emission tomography, with respect to the dispersion of the input function. J Cereb Blood Flow Metab. 1986; 6(5):536–45.View ArticlePubMedGoogle Scholar
- Munk OL, Keiding S, Bass L. A method to estimate dispersion in sampling catheters and to calculate dispersion-free blood time-activity curves. Med Phys. 2008; 35(8):3471–81. http://dx.doi.org/10.1118/1.2948391.
- Kim JC, Anderson SE, Kaye W, Zhang F, Zhu Y, Kaye SJ, He Z. Charge sharing in common-grid pixelated CdZnTe detectors. Nucl Inst Methods Phys Res Sect A Accelerators Spectrometers Detectors Assoc Equip. 2011; 654(1):233–43. iSSN 0168-9002. http://www.sciencedirect.com/science/article/pii/S0168900211011545.
- Boucher YA, Jaworski JM, Kaye WR, Zhang F, He Z. Results from testing of 145 3D position-sensitive, Pixelated CdZnTe Detectors. IEEE Trans Nucl Sci. 2012; 59(6):3332–8. iSSN 0018-9499.View ArticleGoogle Scholar
- Luke P. Unipolar charge sensing with coplanar electrodes-application to semiconductor detectors. IEEE Trans Nucl Sci. 1995; 42(4):207–13.View ArticleGoogle Scholar
- He Z. Review of the Shockley-Ramo theorem and its application in semiconductor gamma-ray detectors. Nucl Inst Methods Phys Res Sect A Accelerators Spectrometers Detectors Assoc Equip. 2001; 463(1–2):250–67. iSSN 0168-9002. http://www.sciencedirect.com/science/article/pii/S0168900201002236.
- Hamel L-A, Julien M. Generalized demonstration of Ramo’s theorem with space charge and polarization effects. Nucl Inst Methods Phys Res Sect A Accelerators Spectrometers Detectors Assoc Equip. 2008; 597(2–3):207–11. iSSN 0168-9002. http://www.sciencedirect.com/science/article/pii/S0168900208014393.
- Luke PN, Eissler EE. Performance of CdZnTe coplanar-grid gamma-ray detectors. IEEE Trans Nucl Sci. 1996; 43(3):1481–6. iSSN 0018-9499.Google Scholar
- Knoll GF. Radiation detection and measurement, 4th Edition. New York: John Wiley and Sons; 2010.Google Scholar
- Blood Sampler twilite two. 2016. http://www.swisstrace.com/specifications.html.
- Roehrbacher F, Bankstahl JP, Bankstahl M, Wanek T, Stanek J, Sauberer M, Muellauer J, Schroettner T, Langer O, Kuntner C. Development and performance test of an online blood sampling system for determination of the arterial input function in rats. EJNMMI Phys. 2015; 2(1):1.View ArticlePubMedPubMed CentralGoogle Scholar